Dentsply-Sirona Sirona Heliodent Plus DC Periapical X-Ray Device
What is the Product HELIODENT Plus (Heliodent Plus) is an intraoral (periapical) X-ray device manufactured by Dentsply Sirona / Sirona. It provides fast and safe intraoral imaging with a DC high-frequency generator and can be used with film, phosphor plates, and digital sensors. It features a 0.4 mm focal spot (IEC 336 / IEC 60336), selectable tube voltage of 60 kV / 70 kV, and a tube current of 7 mA. It uses high voltage generation with a DC high-frequency unit. Usage and Installation Flexibility It adapts to the clinic layout with three different arm lengths and many mounting options. Detector media can be freely selected for transition from film to digital radiography. Arm lengths (manufacturer): 1500 mm (M), 1790 mm (L), 2040 mm (XL); round or square design options Mounting: wall, ceiling, mobile stand, or Dentsply Sirona unit integration Control: remote control, remote timer, or manual trigger Clinical Applications Dental caries diagnosis General horizontal and vertical bone defect assessment Endodontic measurements Implantology — postoperative condition and implant position control Technical Specifications Focal spot size: 0.4 mm (IEC 336) X-ray tube voltage: adjustable between 60 kV and 70 kV X-ray tube current: 7 mA Generator: high frequency DC generator for regulated high voltage Line voltage: 120 V – 240 V, 50–60 Hz Exposure time: steplessly adjustable between 0.01 s and 3.2 s Detector media: film, phosphor plate, or digital sensor (free choice); switching possible between two detector media Standard focus-to-skin distance: 20 cm (200 mm); optional: 30 cm (300 mm) Automatic sleep mode: can be activated after user-defined time