Kikusui PCR6000WEA2 3P4W 400V AC Supply Power NEW
Features: Compact size: 6 kVA in 6U (PCR6000WEA2/PCR6000WEA2R) Up to 36 kVA in a Single Unit AC output *1: Single-phase/Single-phase three-wire/Three-phase DC output: +/-, 100% of rated power 100% Regenerative-power capability Expandable up to three-phase 540 kVA※2 Flexible digital interface: LAN (LXI), USB, RS232C, GPIB (option) Power line disturbance simulation features Sequence function External analog, digital control function (standard) Power-saving function Output frequency up to 5 kHz Output rating: AC 0 to 320 Vrms, DC 0 to ±452 V Product Images Model Phase Power capacity[VA] Phase voltage[V] Max. current (L/H range)[A]※1 Frequency[Hz] PCR1000WEA Single phase 1k (The spec guaranteed voltage range) 1 to 160/ 2 to 320 (L/H output range) (Voltage setting range) 0 to 161.0/ 0 to 322.0 10 / 5 1 to 5000 PCR2000WEA Single phase 2k 20 / 10 PCR3000WEA2 Single phase 3k 30 / 15 Three-phase 10 / 5 Single-phase Three-wire 2k PCR6000WEA2 Single phase 6k 60 / 30 Three-phase 20 / 10 Single-phase Three-wire 4k PCR12000WEA2 Single phase 12k 120 / 60 Three-phase 40 / 20 Single-phase Three-wire 8k PCR18000WEA2 Single phase 18k 180 / 90 Three-phase 60 / 30 Single-phase Three-wire 12k PCR24000WEA2 Single phase 24k 240 / 120 Three-phase 80 / 40 Single-phase Three-wire 16k PCR30000WEA2 Single phase 30k 300 / 150 Three-phase 100 / 50 Single-phase Three-wire 20k PCR36000WEA2 Single phase 36k 360 / 180 Three-phase 120 / 60 Single-phase Three-wire 24k Input Model Single-phase output Single-phase/three-phase switchable model PCR1000WEA PCR2000WEA PCR3000WEA2 PCR6000WEA2 PCR12000WEA2 PCR18000WEA2 PCR24000WEA2 PCR30000WEA2 PCR36000WEA2 PCR6000WEA2R PCR12000WEA2R PCR18000WEA2R PCR24000WEA2R PCR30000WEA2R PCR36000WEA2R Nominal input voltage 1P2W input model 100 Vac to 120 Vac / 200 Vac to 240 Vac *1 – 3P3W 200 V input model / 3P3W 400 V input model – 200 Vac to 240 Vac (3 phase line voltage) *2 / 380 Vac to 480 Vac (3 phase line voltage) *2 *6 3P4W input model – 380 Vac to 480 Vac (3 phase line voltage) *3 Phase Single-phase Three-phase Nominal input Frequency 50 Hz to 60 Hz Input frequency range 45 Hz to 65 Hz Apparent power 1.4 kVA and less 2.7 kVA and less 4 kVA and less 7.8 kVA and less 15.6 kVA and less 23.4 kVA and less 31.2 kVA and less 39 kVA and less 46.8 kVA and less Power factor *4 0.95(TYP) 200 V: 0.97(TYP), 400 V: 0.95(TYP) 3P3W input model / 0.95(TYP) 3P4W input model Maximum current 1P2W input model *1 17 A / 8.5 A 32 A / 16 A 48 A / 24 A – 3P3W 200 V input model – 27 A 53 A 80 A 106 A 133 A 159 A 3P3W 400 V input model 14 A 28 A 42 A 56 A 70 A 84 A 3P4W input model – 14 A 28 A 42 A 56 A 70 A 84 A Hold-up time for power interruption *4 10 ms Protective conductor current *5 1P2W input model 3.5 mA or less – 3P3W 200 V input model – 10 mA or less 15 mA or less 20 mA or less 25 mA or less 30 mA or less 35 mA or less 3P3W 400 V input model – 10 mA or less 20 mA or less 30 mA or less 40 mA or less 50 mA or less 60 mA or less 3P4W input model – 3.5 mA or less *1 100 V/200 V input system (auto select) *2 Models with an “R” in the name (6 kVA or higher) have a three-phase, three-wire input. *3 Models of 6 kVA or more without an “R” in the name have three-phase 4-wire input. *4 At output voltage 100 V/200 V, rated output current, sine wave, load power factor 1, output frequency 40 Hz to 1 kHz *5 At output voltage 100 V/200 V, rated output current, sine wave, load power factor 1, output frequency 45 Hz to 65 Hz *6 For the 400V input model, do not connect to transformer equipment with isolation voltage exceed-ing 300 V. It cannot be connected to a delta-connected power distribution facility with one pole grounded. Output Model Single-phase output Single-phase/three-phase switchable model PCR1000WEA PCR2000WEA PCR3000WEA2 PCR6000WEA2 PCR12000WEA2 PCR18000WEA2 PCR24000WEA2 PCR30000WEA2 PCR36000WEA2 PCR6000WEA2R PCR12000WEA2R PCR18000WEA2R PCR24000WEA2R PCR30000WEA2R PCR36000WEA2R Maximum peak current *11 4 times the maximum output current Inrush current capacity *3 3 times the rated current (0.07 s) *12 1.4 times the rated current (0.5 s) 1.4 times the rated current (0.5 s) Efficiency *10 82 %(TYP) 85 %(TYP) AC Voltage Rating 160 V / 320 V *2 Setting range 0 V to 161.0 V, 0 V to 322.0 V Resolution 0.1 V Setting accuracy (phase voltage) *3 *4 ±(0.3 % of setting + 0.3 V), ±(0.3 % of setting + 0.6 V) Setting accuracy (Line voltage) *3 *4 ±(0.3 % of setting + 0.3 V), ±(0.3 % of setting + 0.6 V) *5 Maximum current *1 *6 Single-phase output 10 A / 5 A 20 A / 10 A 30 A / 15 A 60 A / 30 A 120 A / 60 A 180 A / 90 A 240 A / 120 A 300 A / 150 A 360 A / 180 A Single-phase three-wire output, Three-phase output – 10 A / 5 A 20 A / 10 A 40 A / 20 A 60 A / 30 A 80 A / 40 A 100 A / 50 A 120 A / 60 A Phase 1P 1P2W, 1P3W, 3P4W switchable Power capacity Single-phase output 1 kVA 2 kVA 3 kVA 6 kVA 12 kVA 18 kVA 24 kVA 30 kVA 36 kVA Three-phase output – 3 kVA 6 kVA 12 kVA 18 kVA 24 kVA 30 kVA 36 kVA Single-phase three-wire output – 2 kVA 4 kVA 8 kVA 12 kVA 16 kVA 20 kVA 24 kVA Load power factor 0 to 1 (leading or lagging) Frequency Setting range 1 Hz to 5 kHz *7 (5 kHz -3dB, <40 Hz derating required) Resolution 0.01 Hz(1.00 Hz to 100.0 Hz), 0.1 Hz(100.0 Hz to 1000 Hz), 1 Hz(1000 Hz to 5000 Hz) Accuracy *3 ±0.01 %, Temperature coefficient : ±0.005 %/°C Phase Resolution – 0.01*13 , 0.1° (1 Hz to 500 Hz), 1° (500 Hz to 4 kHz), 2° (4 kHz or more) Accuracy *3 – Within ±(0.4° + fo×0.9°) *8 fo: frequency [kHz] [kHz] DC Voltage voltage Rating *1 -226 V to +226 V, -452 V to +452 V *2 Setting range *1 -227.5 V to +227.5 V, -455.0 V to +455.0 V Resolution 0.1 V Accuracy *9 ±（0.05 % of setting + 0.1 V） Maximum current *6 10 A / 5 A 20 A / 10 A 30 A / 15 A 60 A / 30 A 120 A / 60 A 180 A / 90 A 240 A / 120 A 300 A / 150 A 360 A / 180 A Power capacity 1 kW 2 kW 3 kW 6 kW 12 kW 18 kW 24 kW 30 kW 36 kW 1 output L range, output H range *2 Specification guaranteed voltage range is 1 V to 160 V/ 2 V to 320 V (AC) and 1.4 V to 226 V/ 2.8 V to 452 V (DC) *3 At ambient temperature of 23 °C±5 °C. *4 No load, output frequency 45 Hz to 65 Hz *5 When the phase angle of 120° of each phase. *6 For output phase voltage of 100 Vac to 160 Vac/ 200 Vac to 320 Vac and out-put voltage of 100 Vdc to 226 Vdc/ 200 Vdc to 452 Vdc, output current is reduced with output voltage. When the output frequency is between 1 Hz and 40 Hz, the output current is reduced by the output frequency. The output current is 70 % at 1 Hz. *7 On the 500 Hz limit model, the frequency is limited to 1 Hz to 500.0 Hz for three-phase output. *8 Within ±(0.4° + 2.5 μs×360°×fo×103). The following show the angles obtained by calculating the expression with the specified frequency, within ± 0.5° (at 60 Hz output), within ± 0.8° (at 400 Hz output) *9 With no load at 23°C±5°C. *10 When the output voltage is 100 V or 200 V, the output current is the rated value, the load power factor is 1, and the output frequency is between 40 Hz and 1 kHz. *11 Repeated output is possible when the crest factor is 4. *12 125 Vac/ 250 Vac (output L range/ H range) *13 Waveform bank 0, at 1 Hz to 500 Hz. Output Voltage Stability (Phase Voltage) Model Single-phase output Single-phase/three-phase switchable model PCR1000WEA PCR2000WEA PCR3000WEA2 PCR6000WEA2 PCR12000WEA2 PCR18000WEA2 PCR24000WEA2 PCR30000WEA2 PCR36000WEA2 PCR6000WEA2R PCR12000WEA2R PCR18000WEA2R PCR24000WEA2R PCR30000WEA2R PCR36000WEA2R Line regulation *1 Within ±0.1 % Load regulation *2 Within ±0.1 V/ ±0.2 V(1 Hz to 100 Hz) Within ±0.3 V/ ±0.6 V(100.1 Hz to 500 Hz) Within ±1 V/ ±2 V(500.1 Hz to 1 kHz) Within ±0.2 V/ ±0.4 V(1 Hz to 100 Hz) Within ±0.3 V/ ±0.6 V(100.1 Hz to 500 Hz) Within ±1 V/ ±2 V(500.1 Hz to 1 kHz) Output frequency variation *3 When the output voltage correction function is enabled : Within ±0.3 %(1 Hz to 1 kHz), Within ±10 %(1001 Hz to 5 kHz) When the output voltage correction function is disabled : Within -3 dB(5 kHz) Ripple noise *4 ≤ 0.25 Vrms Ambient temperature variation *5 ±100 ppm/ °C (TYP) Total harmonic distortion *6 0.3 % and less(1 Hz to 100 Hz), 0.5 % and less(100.1 Hz to 330 Hz), 1.5 %/kHz and less(330.1 Hz to 5 kHz) Transient response *7 Response FAST : 40 μs(TYP) Response speed Tr/Tf *8 Response FAST : 40 μs(TYP) Response MEDIUM : 100 μs(TYP) Response SLOW : 300 μs(TYP) * 1 With respect to changes in the rated range of input voltage. 2 With respect to 0 % to 100 % changes in the rating of output current. When the output phase voltage is between 80 V and 160 V (L range) or 160 V and 320 V (H range) and the load power factor is 1, and the response is FAST. At the output terminal block, when the compensation function is not used. * 3 Voltage variation over 40 Hz to 5 kHz in AC mode with 55 Hz as the reference. When the output phase voltage is between 80 V and 160 V or 160 V and 320 V and the load power factor is 1, and the response is FAST, at the output terminal block. * 4 5 Hz to 1 MHz components in DC mode. * 5 With respect to changes in the operating temperature range. When the output phase voltage is 100 V or 200 V, with no load. * 6 When the output phase voltage is between 80 V and 160 V or 160 V and 320 V and the load power factor is 1, and the response is FAST, at the output terminal block. * 7 When the output voltage is 100 V or 200 V, the load power factor is 1, and the output current changes from 0 A to the rated value and from the rated value to 0 A. * 8 At 10 % to 90 % of the output voltage. The PCR6000WEA2 3P4W 400V (500Hz) from Kikusui is an ultra-compact AC/DC power supply with a power rating of 6000 watts. It operates at a voltage of 320 volts and delivers a current of 60 amperes. This model features a single output and supports remote control functionality, allowing for convenient operation. The power supply is designed to work with a three-phase, four-wire 400-volt system and can handle frequencies up to 500 hertz. Its compact design makes it suitable for applications requiring a reliable and efficient power source within a limited space. The combination of high power output and remote control capability provides flexibility for various industrial and testing environments. *The image is for reference only. See product specification for details.