HORIBA Compact Nitrate Ion Meter LAQUAtwin NO3-11
SUMMARY This compact nitrate ion meter measures nitrate ion concentration by applying a small amount of sample, from only 0.3 mL, directly to the flat sensor. LAQUAtwin's unique planar ion-selective electrode enables quick and reliable measurement over a wide concentration range in a compact, portable format. The general-purpose NO3-11 is useful for soil and crop management, food and agricultural product testing, environmental measurement, and other nitrate analysis applications. FEATURES Measures nitrate ion concentration from only 0.3 mL of sample Measurement possible from 0.05 mL when using Sampling Sheet B Flat nitrate ion-selective sensor Measures both nitrate ion (NO3−) and nitrate-nitrogen (NO3-N) Wide nitrate measurement range from 6 to 9900 ppm (mg/L) Up to two calibration points Automatic recognition of the default 150 ppm and 2000 ppm standard solutions Low- and high-concentration calibration points can be adjusted Multiplication compensation from 0.01 to 9.90 Concentration, temperature, and mV display modes Temperature sensor adjustment function Auto Hold function Auto Stability function Backlit monochrome digital LCD Automatic power-off after 30 minutes Remaining battery level display IP67 water-resistant construction Replaceable sensor Portable storage case and measurement accessories included VERSATILE SAMPLING METHODS LAQUAtwin's flat-sensor design allows samples to be applied or collected in several convenient ways. Choose the method that suits the sample and measurement purpose. Note: Depending on the sample, preparation such as extraction, dilution, filtration, or the addition of pure water may be required. Use consistent preparation conditions when comparing measurement results. Direct Drop Measurement Apply a liquid sample directly to the flat sensor with a dropper, syringe, or pipette. The NO3-11 requires at least 0.3 mL for direct measurement. Sampling Sheet B allows measurement from as little as 0.05 mL. Immersion Measurement For liquid samples in a beaker or container, the sensor can be immersed directly in the sample. Open the sliding cap on the sensor guard before measurement. Scoop Sampling The sensor guard can be used to scoop a liquid sample directly from a river, lake, tank, nutrient solution, or other sampling location for convenient on-site measurement. Moist Solid Samples Moist solid samples such as foods or plant materials can be cut into small pieces and placed directly on the flat sensor when sufficient liquid makes contact with the electrode. Powder Samples Place the powder sample on the flat sensor and add a fixed amount of pure water before measurement. Keep the sample amount and water volume consistent when comparing results. Paper, Textiles, and Sheet-like Samples Cut the sample into small pieces, place them on the flat sensor, and add a fixed amount of pure water to measure nitrate extracted from the material. SPECIFICATIONS Model LAQUAtwin-NO3-11 Intended Use General nitrate measurement Measurement Principle Ion-selective electrode method Required Sample Volume 0.3 mL or more Measurement from 0.05 mL is possible when using Sampling Sheet B*1 Measurement Range Nitrate ion (NO3−): 6 to 9900 ppm (mg/L) Nitrate-nitrogen (NO3-N): 1.4 to 2200 ppm (mg/L) Resolution 0 to 99 ppm: 1 ppm 100 to 990 ppm: 10 ppm 1000 to 9900 ppm: 100 ppm Temperature Display Range / Resolution 0 to 50°C / 0.1°C Calibration Up to 2 points Default calibration values: 150 ppm and 2000 ppm Accuracy ±10% of the reading*2 Main Functions Adjustable low- and high-concentration calibration points Multiplication compensation coefficient: 0.01 to 9.90 Concentration, temperature, and mV display modes Temperature sensor adjustment function Auto Hold Auto Stability function Automatic power-off after 30 minutes Remaining battery level display Replaceable sensor Water Resistance IP67 equivalent*3 Display Custom monochrome digital LCD with backlight Operating Environment 5 to 40°C, 85% relative humidity or less, non-condensing Power Supply Two 3 V CR2032 coin-type lithium batteries Battery Life Approximately 400 hours of continuous use with the backlight switched off*4 Main Material ABS resin Dimensions and Weight W164 × H29 × D20 mm, excluding protruding parts Approximately 50 g, excluding batteries Included Accessories S040 nitrate ion sensor Two CR2032 coin-type lithium batteries One dropper Instruction manual Quick manual Storage case 150 ppm nitrate ion standard solution, 14 mL × 1 bottle 2000 ppm nitrate ion standard solution, 14 mL × 1 bottle Sampling Sheet B × 5 sheets Main Consumables S040: Nitrate ion sensor Y043: 2000 ppm nitrate ion standard solution Y045: 150 ppm nitrate ion standard solution *1 Measurement is possible from 0.05 mL when using Sampling Sheet B. Close the light shield cover during measurement. For samples containing fine particles, use a separate sampling-sheet holder cover. *2 Indicates agreement between the measured value and the value of a standard solution after two-point calibration using the supplied standard solutions. Standard solution error and rounding error of ±1 digit are not included. *3 Equivalent to IP67. The meter will not fail after immersion in water at a depth of 1 m for 30 minutes. It cannot be used underwater. *4 Battery life is reduced when the backlight is used. The backlight remains illuminated for approximately 20 seconds. Technical Measurement Notes Nitrate-nitrogen (NO3-N) represents the nitrogen component contained in nitrate. It is not a measurement of total nitrogen. Measurement values may be affected by ions other than nitrate contained in the sample. Iodide, nitrite, bromide, and chloride ions may interfere with nitrate measurement. Samples containing perchlorate ions (ClO4−) cannot be measured with this sensor. Oils, fats, high conductivity, and extreme acidity or alkalinity may also affect measurement results. For quantitative comparisons, keep the sample preparation, dilution or extraction ratio, temperature, calibration procedure, and measurement conditions consistent.