Anaerobic Incubator Internal Volume of Incubator: 339L Incubation Chamber Temperature Range: 15℃ ~ 60℃ Anaerobic Grade: Oxygen content in operation chamber ≤ 0.5% The chamber is equipped with a UV germicidal lamp Application The anaerobic incubator is a specialized device for bacterial culture and operation under anaerobic conditions. It provides a strict anaerobic environment, constant temperature incubation conditions, and a systematic, scientific workspace. It is applied in the testing and research of pathogenic anaerobic bacteria, microbiological inspection for food hygiene, and other fields requiring hypoxic, microaerobic, and anaerobic experimental conditions. Features It consists of a constant temperature incubator, anaerobic operation chamber, sampling chamber, gas and circuit control systems, cabinet stand, bottle rack, and wax melting sterilizer. By using different components of mixed gases, it can create stable extreme anaerobic, microaerophilic, and low-oxygen environments, flexibly cultivating different microbial culture conditions. Equipped with a microcomputer intelligent PID temperature controller, which accurately reflects and controls the temperature inside the incubator. The front windows of the operation and sampling chambers are made of impact-resistant special transparent glass. Dedicated gloves are used for operation, ensuring flexibility, convenience, reliability, and comfort. The operation chamber is equipped with a specialized inoculating loop sterilizer, wax melting disinfection device, and deoxygenation catalyst. The inner chamber of the incubator is made of SUS304 stainless steel. The chamber is equipped with a UV germicidal lamp. Uses a Cubigel refrigeration compressor, high efficiency and low consumption; uses fluorine-free environmentally friendly refrigerant. (Optional) Independent external temperature protection device. (Optional) Two sets of 0.22μm replaceable HEPA high-efficiency filters: Set A is used to remove trace impurities from the injected gas; Set B is used to remove harmful microorganisms and odors from the exhaust gas. (Based on actual usage, the recommended filter replacement interval is 3 to 6 months.) Specifications Model ICB-AN-HYQIII-Z Incubation Chamber Temperature Range 15℃ ~ 60℃ Temperature Uniformity ≤±1℃ Temperature Fluctuation ≤ ±0.3℃ Anaerobic Grade Oxygen content in operation chamber ≤ 0.5% Anaerobic State Formation Time (Sampling Chamber) ≤ 15 minutes Anaerobic (Microaerophilic) Formation Method (Sampling Chamber) Vacuum + Gas displacement (using 99.99% N₂ or mixed gas of 5% H₂, 10% CO₂, and 85% N₂) Vacuum + Gas displacement (using 99.99% N₂ or mixed gas of 5% O₂, 10% CO₂, and 85% N₂) Anaerobic Process Operation (Sampling Chamber) One-touch automatic execution Anaerobic State Formation Time (Operation Chamber) ≤ 1.5 hours Anaerobic (Microaerophilic) Formation Method (Operation Chamber) Vacuum + Gas displacement (using 99.99% N₂ or mixed gas of 5% H₂, 10% CO₂, and 85% N₂) Vacuum + Gas displacement (using 99.99% N₂ or mixed gas of 5% O₂, 10% CO₂, and 85% N₂) Anaerobic Maintenance Time (Operation Chamber) > 12 hours (after stopping trace mixed gas supplementation) Anaerobic Process Operation (Operation Chamber) Manual button switching Internal Dimensions of Incubation Chamber Operation chamber/Incubation chamber Internal Volume of Incubator 339L Sampling Chamber Dimensions 40×33×32cm Operation Chamber Dimensions 85.2×56.9×70cm External Dimensions 140×68.8×141.8cm Temperature Controller LCD display, PID temperature controller Shelves 2 Electricity 1Ø 220V, 50Hz Power 1000W Note: The external height (H) includes 8cm accessories. Customers are required to provide their own experimental gas cylinders, gases, and pressure reducing valves.