AC High-Capacity Freezer 430 Tight Neck Opening - MDR
#1LEVEL-WITHAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. RACK STORAGE SYSTEM FOR AC BIO 430 Inventory system for storage of cryogenic vials. Complementary for container AC BIO 430. It consists of: 12x racks for 10x10 cryoboxes 4x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 16.900 pcs. Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Rack for 10x10 cryoboxes - H735 x W138 x D136 mm Racks for 5x5 cryoboxes - H735 x W82 x D76 mm #1LEVEL-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (Rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC BIO 430 Inventory system for storage of cryogenic straws or vials. Complementary for container AC BIO 430. It consists of: 76x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Storage capacity: Straws 0,25ml: 117.800 pcs. Straws 0,5ml: 52.440 pcs. Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #1LEVEL-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (Rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #1LEVEL-WITHOUTAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (Rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. RACK STORAGE SYSTEM FOR AC BIO 430 Inventory system for storage of cryogenic vials. Complementary for container AC BIO 430. It consists of: 12x racks for 10x10 cryoboxes 4x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 16.900 pcs. Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Rack for 10x10 cryoboxes - H735 x W138 x D136 mm Racks for 5x5 cryoboxes - H735 x W82 x D76 mm #1LEVEL-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (Rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC BIO 430 Inventory system for storage of cryogenic straws or vials. Complementary for container AC BIO 430. It consists of: 76x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Storage capacity: Straws 0,25ml: 117.800 pcs. Straws 0,5ml: 52.440 pcs. Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #1LEVEL-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC BIO 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC BIO 430 Part No. AC BIO 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 750 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 195 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 1 Number of racks 12x of 10x10 Cryoboxes / 4x of 5x5 Cryoboxes Straw holding capacity 0,25 ml 117.800 Straw holding capacity 0,5 ml 52.440 2ml Vials capacity (Rack) 16.900 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: #2LEVELS-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC IVF 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC IVF 430 Part No. AC IVF 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 2x 370 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 197 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 2 Straw holding capacity 0,25 ml 123.690 Straw holding capacity 0,5 ml 55.195 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. two-storey carousel with box to accommodate the piping and valves without storage system no MDD Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC IVF 430 Inventory system for storage of cryogenic straws in 65 mm goblets or vials in vial holders. (goblets and vial holders are not included). Complementary for container AC IVF 430. It consists of: 133x stainless steel canisters 370 mm x 70 mm (height x diameter). Storage capacity: Straws 0,25ml: 123.690 pcs. Straws 0,5ml: 55.195 pcs. Note: The product image is a sample illustration. #2LEVELS-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC IVF 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC IVF 430 Part No. AC IVF 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 2x 370 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 197 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 2 Straw holding capacity 0,25 ml 123.690 Straw holding capacity 0,5 ml 55.195 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. two-storey carousel with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #2LEVELS-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC IVF 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC IVF 430 Part No. AC IVF 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 2x 370 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 197 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 2 Straw holding capacity 0,25 ml 123.690 Straw holding capacity 0,5 ml 55.195 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. two-storey carousel with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC IVF 430 Inventory system for storage of cryogenic straws in 65 mm goblets or vials in vial holders. (goblets and vial holders are not included). Complementary for container AC IVF 430. It consists of: 133x stainless steel canisters 370 mm x 70 mm (height x diameter). Storage capacity: Straws 0,25ml: 123.690 pcs. Straws 0,5ml: 55.195 pcs. Note: The product image is a sample illustration. #2LEVELS-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC IVF 430 Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running aluminum storage system on one level for the AC BIO series and two levels for the AC IVF series. The containers of both series are characterized by very low nitrogen consumption at a stable storage temperature, which almost reaches the cold level of liquid nitrogen even in the gas phase. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family Freezer Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC IVF 430 Part No. AC IVF 430 LN2 capacity (liters) 430 LN2 fill height [mm] 135 LN2 value below the base plate [l] 40 Outside diameter (mm) 800 Total height (mm) 1,342 Usable height [mm] 2x 370 Neck opening (mm) 317 Empty weight (kg) (without ICS**) 197 Static evaporation rate (liters / day) 9 Static hold time (days) Number of castors 5 Number of storage carousels 2 Straw holding capacity 0,25 ml 123.690 Straw holding capacity 0,5 ml 55.195 * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. ** ICS = Inventory Control System Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. two-storey carousel with box to accommodate the piping and valves without storage system MDR Function: The cooling effect required for cryogenic sample storage is generated by the continuous evaporation of the liquid nitrogen refrigerant. The necessary heat of vaporization is extracted from the environment by heat transfer. As a result, the liquid nitrogen level in the deep-freeze storage tanks is constantly falling. Regular, controlled refilling is therefore required (automatically or manually). This results in the liquid level curve shown here. In order to prevent unnecessary nitrogen losses and icing of the storage containers, liquid nitrogen should be refilled at pressures between 0.7 bar - 1.4 bar only. The insulating properties of the Auguste Cryogenics Medical Family are essentially based on the presence of a vacuum between the inner and outer container. A vacuum loss inevitably leads to a very high increase in nitrogen consumption and poses a risk for the stored samples if there is no refill with liquid nitrogen. Download data sheet & technical drawing: #searchwords biobank storage tank biobank freezers high capacity cryo tank cryogenic storage tank biobank storage system vapor phase storage tank liquid phase ln2 storage biorepository cryogenic freezer cryogenic storage vessel cryogenic storage system cryogenic preservation system cryogenic archive system low temperature storage system long term sample storage biospecimen storage biobank storage system nitrogen vapor phase storage liquid phase cryostorage high capacity cryostorage automated cryostorage system sample archive system cryogenic dewar sample storage dewar cell banking system cryostorage for cell lines mammalian cell storage cell cryostorage cell line repository stem cell cryostorage stem cell biobank dna biobank rna sample storage biomaterial storage cell banking cryopreservation high density cryogenic storage platform biobankstoragetank biobankfreezers highcapacitycryotank cryogenicstoragetank biobankstoragesystem vaporphasestoragetank liquidphaseln2storage biorepository cryogenicfreezer cryogenicstoragevessel cryogenicstoragesystem cryogenicpreservationsystem cryogenicarchivesystem lowtemperaturestoragesystem longtermsamplestorage biospecimenstorage biobankstoragesystem nitrogenvaporphasestorage liquidphasecryostorage highcapacitycryostorage automatedcryostoragesystem samplearchivesystem cryogenicdewar samplestoragedewar cellbankingsystem cryostorageforcelllines mammaliancellstorage cellcryostorage celllinerepository stemcellcryostorage stemcellbiobank dnabiobank rnasamplestorage biomaterialstorage cellbanking cryopreservation highdensitycryogenicstorageplatform biobank storage tank biobank freezer hochkapazitäts kryotank kryogener lagertank biobank lagersystem dampfphasen lagertank flüssigphasen ln2 lagerung biorepositorium kryolagerbehälter kryotank kryobehälter flüssigstickstoff lagerbehälter ln2 lagertank kryolagersystem kryokonservierungssystem kryospeichersystem tieftemperatur lagersystem langzeit probenlager bioprobenlager biobank lagersystem stickstoff dampfphasenlager flüssigphasen kryolager hochkapazitäts kryolager automatisiertes kryolagersystem probenarchivierungssystem biorepositorium kryodewar dewar gefäß probenlagerdewar cell banking system zellbank lagersystem kryolager für zelllinien lagerung von säugerzellen zell kryolager zelllinien repository stammzell kryolager stammzell biobank dna biobank rna probenlager biomateriallager cell banking kryokonservierung hochdichte kryogene lagerplattform biobankstoragetank biobankfreezer hochkapazitätskryotank kryogenerlagertank biobanklagersystem dampfphasenlagertank flüssigphasenln2lagerung biorepositorium kryolagerbehälter kryotank kryobehälter flüssigstickstofflagerbehälter ln2lagertank kryolagersystem kryokonservierungssystem kryospeichersystem tieftemperaturlagersystem langzeitprobenlager bioprobenlager biobanklagersystem stickstoffdampfphasenlager flüssigphasenkryolager hochkapazitätskryolager automatisierteskryolagersystem probenarchivierungssystem biorepositorium kryodewar dewargefäß probenlagerdewar cellbankingsystem zellbanklagersystem kryolagerfürzelllinien lagerungvonsäugerzellen zellkryolager zelllinienrepository stammzellkryolager stammzellbiobank dnabiobank rnaprobenlager biomateriallager cellbanking kryokonservierung hochdichtekryogenelagerplattform Specifications LN2 capacity in gas storage (liters): 40 LN2 capacity in liquid storage (liters): 430 Empty weight (kg): 210 Neck opening (mm): 317 Outside diameter (mm): 800 Total height (mm): 1419
AC BIO 430 - MDR + CryoFill 2.0 PD MDR with Installation + Rack Storage System for AC BIO 430
2. Please choose: AUTOMATIC FILLING SYSTEM
WITH RACK STORAGE SYSTEM, WITH CANISTER STORAGE SYSTEM, WITHOUT STORAGE SYSTEM
AC BIO 430 - MDR + CryoFill 2.0 PD MDR with Installation + Canister Storage System for AC BIO 430
AC BIO 430 - MDR + CryoFill 2.0 PD MDR with Installation
AC BIO 430 - MDR + Rack Storage System for AC BIO 430
AC BIO 430 - MDR + Canister Storage System for AC BIO 430
AC IVF 430 - MDR + CryoFill 2.0 PD MDR with Installation + Canister Storage System for AC IVF 430
AC IVF 430 - MDR + CryoFill 2.0 PD MDR with Installation
AC IVF 430 - MDR + Canister Storage System for AC IVF 430