KUAIQU SPPS-Q605 60V 5A 300W High-Precision DC Power Supply for Industrial Electronics
KUAIQU SPPS-Q605 60V 5A 300W High-Precision DC Power Supply Product Overview The KUAIQU SPPS-Q605 60V 5A DC Power Supply is a wide-voltage bench power supply designed for 48V systems, industrial electronics, communication equipment, power modules, and higher-voltage DC testing. Its adjustable 0–60V output provides greater voltage flexibility than a conventional 30V power supply, while the 5A current capability and 300W maximum power support a wide range of electronic loads. The SPPS-Q605 combines a 60V adjustable DC power supply with a color LCD display, USB-A/USB-C fast charging, real-time USB monitoring, and OCP, OVP, and OPP protection. The wider voltage range allows users to test different DC voltage levels, including common 24V, 36V, and 48V systems. Key Features 0–60V / 0–5A Adjustable Output: Provides up to 300W for 48V equipment, industrial electronic modules, communication devices, power electronics, and other DC loads. Wide 0–60V Voltage Range: Provides flexibility for testing 24V, 36V, 48V, and other DC systems requiring an adjustable output up to 60V. Color LCD Display: Clearly displays voltage, current, and power in real time for convenient monitoring during industrial electronics testing. Precision Low-Voltage Display: Below 10V, output voltage is displayed to three decimal places for detailed low-voltage circuit adjustment and testing. OCP, OVP & OPP Protection: Over-current, over-voltage, and over-power protection helps protect connected equipment during laboratory and industrial testing. USB-A & USB-C Fast Charging: Dual USB ports provide up to 18W output with automatic protocol adaptation for compatible USB devices. Real-Time USB Monitoring: USB voltage and current are displayed in real time for convenient charging and USB device evaluation. Automatic A/mA & W/mW Conversion: Automatically switches measurement units according to the measured value for easier monitoring across different load levels. Output ON/OFF & Panel LOCK: Provides convenient output control and helps prevent accidental panel operation during testing. Buzzer Alarm with Mute: Audible notifications indicate operating or protection events, while the buzzer can be muted when silent operation is required. Applications 48V System Testing: Suitable for 48V electronic equipment, power modules, and DC systems. Communication Equipment: Suitable for 48V-class communication and networking equipment requiring adjustable DC power. Industrial Electronics Testing: Suitable for industrial control modules, electronic controllers, and DC-powered industrial equipment. 24V & 36V Equipment Testing: The 0–60V output range provides flexibility for testing equipment operating at different DC voltage levels. DC-DC Converter Testing: Suitable for evaluating DC-DC converters and power conversion modules across a wider input voltage range. Motor & Drive Electronics: Suitable for testing motor-related electronic modules and drive circuits within the power supply's voltage, current, and power capacity. Power Electronics Testing: Suitable for power modules and electronic circuits requiring adjustable DC input up to 60V. The SPPS-Q605 is a flexible 60V 5A bench power supply for 48V systems and industrial electronics, particularly suited to applications requiring a wider adjustable voltage range and up to 300W of DC output power. Quality Check and Terms of Service Has passed CE & FCC Certification. All DC power supplies and accessories are inspected before shipment to ensure good quality. Customer support: Contact us with any questions, and we'll respond within 24 hours. Package includes: DC Power Supply *1 Output Power Cord *1 Input Power Cord *1 User Manual *1 Warranty Card *1 Product Parameters Parameter Item SPPS-Q605 Rated Values Voltage 0~60V Current 0~5A Power 300W Load Regulation Voltage ≤ 0.01%+25mV Current ≤ 0.1%+15mA Line Regulation Voltage ≤0.01%+15mV Current ≤0.1%+5mA Setting Resolution Voltage <10V: 1mV ≥10V: 10mV Current 1 mA Readback Resolution (Display Resolution) Voltage <10V: 1mV ≥10V: 10mV Current < 20mA: 10 uA ≥ 20mA: 1 mA Setting Accuracy Voltage ≤0.02%+30mV Current ≤0.1%+15mA Readback Accuracy (Display Accuracy) Voltage ≤0.02%+30mV Current <20mA: 0.1%+100uA 20mA~5A: ≤0.01%+15mA Ripple (20Hz～20MHz) Voltage ≤100mV(p-p) ≤25mV(rms) Current ≤50mA(rms) Rise Time (No Load) Voltage ≤100ms Rise Time (Full Load) Voltage ≤1200ms Fall Time (No Load) Voltage ≤10000ms Fall Time (Full Load) Voltage ≤100ms Efficiency (Full Power) / ≥83% Input Voltage Range Vac 200Vac~240Vac Minimum Display Current mA 0.01mA Fuse 220v Input 3.15A Product Dimensions / 218mm89mm150mm Product Weight / 1.2KG