AccuTherm SmartLOG 2021 Digital Data Logger
AccuTherm SmartLOG 2021 Digital Data Logger The AccuTherm SmartLOG 2021 Digital Data Logger (DDL) provides convenient wireless temperature monitoring with real-time email and SMS alerts. Designed for reliable monitoring of temperature-sensitive products, the SmartLOG 2021 includes one wireless temperature sensor with glycol buffer and a 2-year NIST-traceable ISO calibration at 4°C. With Wi-Fi connectivity and configurable logging intervals, the SmartLOG 2021 allows users to monitor conditions and receive alerts without requiring additional software or a cloud subscription. It is CDC/VFC compliant, making it an ideal solution for vaccine storage, pharmacies, laboratories, medical facilities, and other temperature-controlled environments. The AccuTherm SmartLOG 2021 DDL supports up to 2 wireless sensors. Additional AccuTherm Wireless Temperature Sensors are available for purchase separately through the link at the bottom of the page. Key Features Feature Specification Temperature Range -50°C to 70°C Accuracy ±0.5°C (±1.0°F) Probe Type Single wireless temperature sensor with glycol buffer Smart Probe Enabled Yes Smart Probe ACCSLBLET Primary Power USB +5V wall adapter Secondary Power AAA battery Logging Interval Configurable from 1 to 60 minutes Audible Alarm Yes Visual Alarm Yes Email / SMS Alerts Yes NIST-Traceable ISO Calibration Certificate Yes – 2-year calibration at 4°C Report Types PDF, CSV, and graph DDL Type Wi-Fi Software Required No Cloud Subscription Required No Network Requirements Customer Wi-Fi network; password protected CDC / VFC Compliant Yes