Philips Trilogy Evo Portable hospital-to-home ventilator (Mfd. 2020) Software version 1.06.10.00 with Accessories - Like New
Philips Trilogy Evo Portable Hospital to Home Ventilator Ventilator for Reliable Portable and Long Term Respiratory Support The Philips Trilogy Evo ventilator is a next generation portable life support device designed to deliver consistent, high performance ventilation across hospital, transport and home care environments. It is suitable for both invasive and non invasive ventilation, supporting adult and paediatric patients through the entire care pathway. Engineered with advanced monitoring and adaptive algorithms, Trilogy Evo automatically adjusts to changes in patient condition and leak dynamics. It offers a wide range of ventilation modes, including pressure control, volume control and spontaneous timed ventilation. With an internal battery and detachable battery providing up to 15 hours of operation and an intuitive touchscreen interface, the system supports flexible, mobile and safe respiratory care in all settings. At a Glance Brand Philips Software Version 1.06.10.00 Manufacture Year 2020 Key Ventilation Modes A/C VC and A/C PC (assist control volume and pressure). CPAP and S/T (spontaneous timed ventilation). SIMV (VC and PC), MPV (mouthpiece), PSV and EPAP. Advanced support including AVAPS, Auto Trak and dual limb circuit configurations. Included Accessories Tubing (patient circuit) as shown in product images. Other accessories may vary by configuration and order. For Use in Dynamic Environments With expanded cross functionality, Philips Trilogy Evo is suitable for chronic and critical patients in a variety of settings, including sub acute or chronic care units, patients homes and during everyday activities. The ventilator is designed with robust housing and accessories to protect against damage during travel and medical transport. Replacement Parts & Compatibility (not included) Detachable Battery Part Number: 1127889. Internal Battery Part Number: 1127881. Compatibility: These batteries are designed specifically for the Trilogy Evo series, including the Trilogy Evo O2 and EV300 models. Long Lasting Battery Life Ventilator dependent patients are supported by both internal and detachable lithium ion batteries, with a combined nominal run time of up to 15 hours. Compared with older systems such as Trilogy 100, Trilogy Evo offers a higher level of freedom and mobility. It can be mounted on a roll stand or wheelchair and used with a carry bag for transport. Meeting Changing Patient Needs Trilogy Evo provides invasive and non invasive support with high sensitivity for a broad range of adult and paediatric patients. Volume and pressure modes, AVAPS AE, plus optional SpO₂ and EtCO₂ monitoring with alarms on all monitored parameters, allow adaptable care. Flexible circuit options support use across many clinical scenarios. Streamlined Data Accessibility Trilogy Evo uses Bluetooth connectivity to transmit patient and device information to Care Orchestrator, the Philips cloud based management platform. This integrates technologies, data and workflows to support management of respiratory patients. Critical data can also be downloaded directly at the point of care via USB for local review and documentation. Technical Specifications 1. Ventilation Modes A/C PC: assist control (pressure control). A/C VC: assist control (volume control). CPAP: continuous positive airway pressure. PSV: pressure support ventilation. S/T: spontaneous timed ventilation. SIMV PC: synchronous intermittent mandatory ventilation, pressure control. SIMV VC: synchronous intermittent mandatory ventilation, volume control. MPV PC: mouthpiece ventilation, pressure control. MPV VC: mouthpiece ventilation, volume control. 2. Physical Characteristics Weight: 5.2 kg. Size: 16.5 cm D × 28.6 cm W × 24.5 cm H (6.48 in D × 11.25 in W × 9.65 in H). Screen dimensions: 8 in (20.32 cm). 3. Oxygen Low flow: 0 to 30 l/min, maximum 10 psi. High pressure inlet: 280 to 600 kPa (41 to 87 psi). 4. Measured and Displayed Patient Parameters Tidal volume (Vti or Vte): 0 to 2000 ml. Minute ventilation (MinVent): 0 to 30 l/min. Leak: 0 to 200 l/min. Respiratory rate (RR): 0 to 90 breaths per minute. Peak inspiratory flow (PIF): 0 to 200 l/min. Peak inspiratory pressure (PIP): 0 to 90 cm H₂O. Mean airway pressure: 0 to 90 cm H₂O. Percentage spontaneous triggered breaths: 0 to 100 percent. I:E ratio: 9.9:1 to 1:9.9. Dynamic compliance (Dyn C): 1 to 100 ml/cm H₂O. Dynamic resistance (Dyn R): 5 to 200 cm H₂O/l/sec. Dynamic plateau pressure (Dyn Pplat): 0 to 90 cm H₂O. Auto PEEP: 0 to 20 cm H₂O. FiO₂ with FiO₂ sensor: 21 to 100 percent. SpO₂ with pulse oximeter accessory: 0 to 100 percent. Pulse rate with pulse oximeter accessory: 18 to 321 beats per minute. EtCO₂ with CO₂ accessory: 0 to 150 mmHg. 5. Electrical AC input voltage: 100 to 240 V, 50/60 Hz, 1.7 to 0.6 A. DC input voltage: 12/24 V, 6.5 A. Internal and detachable Li ion batteries: 15 hours nominal total run time in accordance with IEC 80601 2 72 (approximately 7.5 hours per battery). Charge time: 0 to 80 percent in about 2.5 hours, 0 to 100 percent in about 3.5 hours. 6. Alarms Inspiratory pressure: PEEP + 1 to 89 cm H₂O. Tidal volume: off, 10 to 2000 ml. Minute ventilation: off, 0.2 to 30 l/min. Respiratory rate: off, 1 to 90 breaths per minute. Circuit disconnection: off, 5 to 60 seconds. Apnoea interval: 5 to 60 seconds. No trigger (MPV only): off, 0.5 to 15 minutes. 7. Controls AVAPS with passive circuit: available in PSV, S/T and A/C PC modes. Tidal volume set range: 35 to 2000 ml (dual limb and active flow circuits); 50 to 2000 ml (passive and active PAP circuits). Breath rate: 0 to 80 breaths per minute. PEEP: 0 to 35 cm H₂O (active circuits); 3 to 25 cm H₂O (passive circuits). EPAP/CPAP: 3 to 25 cm H₂O. IPAP: 3 to 60 cm H₂O. Pressure support or pressure control: 0 to 60 cm H₂O. Inspiratory time: 0.3 to 5.0 seconds. Rise time: 0 to 6. Triggering and cycling: Off, AutoTrak, Sensitive, Flow Trigger. Flow trigger sensitivity: 0.5 to 9 l/min. Flow cycle sensitivity: 10 to 90 percent of peak flow. Flow pattern: square or ramp. FiO₂ set range: 21 to 100 percent. Inspiratory time minimum and maximum: 0.3 to 3.0 seconds. Backup ventilation: on or off. Documentation Trilogy Evo Bench Study Whitepaper (PDF, 384.4 kB) Trilogy Evo Accessory Guide Global (PDF, 12.7 MB)