AC High-Capacity Freezer 480 Wide Neck Opening - MDR
#VIALS2mlINLIQUIDPHASE-WITHAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-2 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 12 2ml Vials capacity 20400 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. RACK STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic vials. Complementary for container AC SYS 480 with storage frame for liquid phase storage. It consists of: 15x racks for 10x10 cryoboxes 8x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 22.100 pcs. (only 20.400 usable in liquid phase) Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors (only 12 floors usable in liquid phase) Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Rack for 10x10 cryoboxes - H735 x W138 x D136 mm Rack for 5x5 cryoboxes - H735 x W82 x D76 mm #VIALS2mlINLIQUIDPHASE-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 without storage frame for liquid phase storage. It consists of: 85x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets with lid, 70 mm diameter Dimension: 715 mm x 75 mm (height x diameter) Note: The product image is a sample illustration. #VIALS2mlINLIQUIDPHASE-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-2 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 12 2ml Vials capacity 20400 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #VIALS2mlINLIQUIDPHASE-WITHOUTAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-2 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 12 2ml Vials capacity 20400 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. RACK STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic vials. Complementary for container AC SYS 480 with storage frame for liquid phase storage. It consists of: 15x racks for 10x10 cryoboxes 8x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 22.100 pcs. (only 20.400 usable in liquid phase) Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors (only 12 floors usable in liquid phase) Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Rack for 10x10 cryoboxes - H735 x W138 x D136 mm Rack for 5x5 cryoboxes - H735 x W82 x D76 mm #VIALS2mlINLIQUIDPHASE-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 without storage frame for liquid phase storage. It consists of: 85x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets with lid, 70 mm diameter Dimension: 715 mm x 75 mm (height x diameter) Note: The product image is a sample illustration. #VIALS2mlINLIQUIDPHASE-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-2 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 12 2ml Vials capacity 20400 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: #VIALS2mlINGASPHASE-WITHAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. RACK STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 15x racks for 10x10 cryoboxes 8x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 22.100 pcs. Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Racks for 10x10 cryoboxes - H735 x W138 x D136 mm Racks for 5x5 cryoboxes - H735 x W82 x D76 mm #VIALS2mlINGASPHASE-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 68x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #VIALS2mlINGASPHASE-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #VIALS2mlINGASPHASE-WITHOUTAUTOFILLING-WITHRACKSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. RACK STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 15x racks for 10x10 cryoboxes 8x racks for 5x5 cryoboxes for storage of 2ml cryogenic vials. (cryoboxes are not included) Maximum storage capacity of 2ml cryo vials: 22.100 pcs. Key features of rack for 10x10 and 5x5 cryoboxes: 13 floors Aluminum 1 mm Locking by means of spring retainer, "snap lock" on each floor Handle Dimensions: Racks for 10x10 cryoboxes - H735 x W138 x D136 mm Racks for 5x5 cryoboxes - H735 x W82 x D76 mm #VIALS2mlINGASPHASE-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 68x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #VIALS2mlINGASPHASE-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: #STRAWSINLIQUIDPHASE-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 without storage frame for liquid phase storage. It consists of: 85x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets with lid, 70 mm diameter Dimension: 715 mm x 75 mm (height x diameter) Note: The product image is a sample illustration. #STRAWSINLIQUIDPHASE-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #STRAWSINLIQUIDPHASE-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 without storage frame for liquid phase storage. It consists of: 85x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets with lid, 70 mm diameter Dimension: 715 mm x 75 mm (height x diameter) Note: The product image is a sample illustration. #STRAWSINLIQUIDPHASE-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITHOUT FRAME FOR LIQUID PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100529 LN2 capacity (liters) 480 Outside diameter (mm) 900 Total height (mm) 1205 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves without storage frame without storage system Download technical drawing: #STRAWSINGASPHASE-WITHAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: 3. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 68x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #STRAWSINGASPHASE-WITHAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CRYOFILL 2.0 PD WITH INSTALLATION ON A DEWAR, WITH MDR APPROVAL (MEDICAL DEVICE) The CryoFill 2.0 is the next generation liquid nitrogen level controller with a number of improved and new features. As well as the ability to have a real time remote read out of relevant data via an alarming and/or monitoring system. New features: Larger, better and more user friendly 4.3” colour touchscreen display with a resolutions of 480x320 Real-time remote read out, compatible with several monitoring systems One Fill All Fill filling sequence (master slave configuration) High temperature control New: One Fill All Fill (OFAF) This is a feature for optimizing the filling sequence of multiple cryogenic storage vessels. The controllers are connected with each other using communication cables. The master controller will make sure all the vessels fill in succession, in order to minimize nitrogen consumption. New: High temperature control This is a new software feature for controlling the temperature in the neck of the vessel. This feature in intended for situation when work in performed in the neck and the lid is open for a longer period. In case the temperature in the neck starts to rise, the controller injects nitrogen into the vessel, which lowers the temperature in the top of the vessel. The CryoFill 2.0 can be used for controlling the liquid nitrogen level for any storage type, either conventional gas phase or the so called “dry” gas phase, with an absolute separation between the samples and liquid nitrogen, or full liquid phase. The CryoFill 2.0 can easily retrofitted to existing cryogenic storage vessels of any brand or type. The CryoFill 2.0 is equipped with the following features and options: User friendly interface Compatible with two measuring standards (pressure differential and 4 point measurement) Two temperature sensors Defog and quick chill functions Hot gas bypass Safety valve, acts as a secondary fill and bypass valve. Unit can be either build onto a vessel or into a cabinet (TW K-series) or into a lid (CBS V- series) Timed filling; Daily fill, weekly fill or fill pulse from cryobank control system Data logging UPS battery back-up Remote real time read out of relevant data via RS485 / RS422 Note: The product image is a sample illustration. Download data sheet: #STRAWSINGASPHASE-WITHOUTAUTOFILLING-WITHCANISTERSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: 2. CANISTER STORAGE SYSTEM FOR AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Inventory system for storage of cryogenic straws or vials. Complementary for container AC SYS 480 with storage frame for gas phase storage. It consists of: 68x aluminium canisters for storing cryo straws in goblets or vials in vial holders. (goblets and vial holders are not included) Key features of aluminium canisters: Contains one lifter Has a perforated base Holds up to 5 goblets without lid, 64 mm diameter Dimension: 715 mm x 70 mm (height x diameter) Caution: This item is not suitable for goblets with lids Note: The product image is a sample illustration. #STRAWSINGASPHASE-WITHOUTAUTOFILLING-WITHOUTSTORAGESYSTEM Consists of: 1. AC SYS 480 WITH FRAME FOR GAS PHASE STORAGE Auguste Cryogenics developed the stainless steel containers in this series from the ground up using state-of-the-art technology. They are specially designed for the safe and economical long-term storage of larger sample quantities and are therefore used in the fields of medicine, pharmaceuticals and research. The vacuum-insulated products come with Auguste Cryogenics' GOLD-standard vacuum and offer a remarkably high storage capacity in a clear, smooth-running stainless steel storage system. The lid of the container opens easily and wide, allowing easy access to the contents. It is convenient to clean and is equipped with a control switch that, in addition to an "open lid" detection, can also send other control signals, for example for automatic demisting or rapid cooling. Auguste Cryogenics Medical Family SYS Series vessels can be supplied with automatic filling and control systems to ensure a regulated supply of cryogenic liquid nitrogen and continuous monitoring. MDR – Medical Device Regulation (EU) The Medical Device Regulation (MDR) defines harmonized requirements across Europe for products intended for medical applications. MDR-compliant nitrogen containers meet the highest standards in terms of safety, quality, and documentation, and are specifically designed for use in sensitive environments such as medical, laboratory, and research settings. They offer verified reliability, compliant design, and full traceability throughout the entire product lifecycle. Technical Data AC SYS 480 Part No. H-ASE1100526-1 LN2 capacity (liters) 480 LN2 volume with storage frame (liters) 90 Outside diameter (mm) 900 Total height (mm) 1205 Space for racks 15x of 10x10 Cryoboxes / 8x of 5x5 Cryoboxes Dimensions of the racks (mm) 145x145 / 87x87 Floors per rack 13 2ml Vials capacity 22100 available as an MDR product yes * The values given are nominal values that were determined without storage system equipment and under precisely defined conditions. Depending on the operating conditions and the storage system equipment used, the actual values can deviate considerably. Due to our policy of continuous product improvement, we reserve the right to change specifications at any time without prior notice. with box to accommodate the piping and valves with storage frame without storage system Download technical drawing: #searchwords biobank storage tank biobank freezers high capacity cryo tank cryogenic storage tank biobank storage system vapor phase storage tank liquid phase ln2 storage biorepository cryogenic freezer cryogenic storage vessel cryogenic storage system cryogenic preservation system cryogenic archive system low temperature storage system long term sample storage biospecimen storage biobank storage system nitrogen vapor phase storage liquid phase cryostorage high capacity cryostorage automated cryostorage system sample archive system cryogenic dewar sample storage dewar cell banking system cryostorage for cell lines mammalian cell storage cell cryostorage cell line repository stem cell cryostorage stem cell biobank dna biobank rna sample storage biomaterial storage cell banking cryopreservation high density cryogenic storage platform biobankstoragetank biobankfreezers highcapacitycryotank cryogenicstoragetank biobankstoragesystem vaporphasestoragetank liquidphaseln2storage biorepository cryogenicfreezer cryogenicstoragevessel cryogenicstoragesystem cryogenicpreservationsystem cryogenicarchivesystem lowtemperaturestoragesystem longtermsamplestorage biospecimenstorage biobankstoragesystem nitrogenvaporphasestorage liquidphasecryostorage highcapacitycryostorage automatedcryostoragesystem samplearchivesystem cryogenicdewar samplestoragedewar cellbankingsystem cryostorageforcelllines mammaliancellstorage cellcryostorage celllinerepository stemcellcryostorage stemcellbiobank dnabiobank rnasamplestorage biomaterialstorage cellbanking cryopreservation highdensitycryogenicstorageplatform biobank storage tank biobank freezer hochkapazitäts kryotank kryogener lagertank biobank lagersystem dampfphasen lagertank flüssigphasen ln2 lagerung biorepositorium kryolagerbehälter kryotank kryobehälter flüssigstickstoff lagerbehälter ln2 lagertank kryolagersystem kryokonservierungssystem kryospeichersystem tieftemperatur lagersystem langzeit probenlager bioprobenlager biobank lagersystem stickstoff dampfphasenlager flüssigphasen kryolager hochkapazitäts kryolager automatisiertes kryolagersystem probenarchivierungssystem biorepositorium kryodewar dewar gefäß probenlagerdewar cell banking system zellbank lagersystem kryolager für zelllinien lagerung von säugerzellen zell kryolager zelllinien repository stammzell kryolager stammzell biobank dna biobank rna probenlager biomateriallager cell banking kryokonservierung hochdichte kryogene lagerplattform biobankstoragetank biobankfreezer hochkapazitätskryotank kryogenerlagertank biobanklagersystem dampfphasenlagertank flüssigphasenln2lagerung biorepositorium kryolagerbehälter kryotank kryobehälter flüssigstickstofflagerbehälter ln2lagertank kryolagersystem kryokonservierungssystem kryospeichersystem tieftemperaturlagersystem langzeitprobenlager bioprobenlager biobanklagersystem stickstoffdampfphasenlager flüssigphasenkryolager hochkapazitätskryolager automatisierteskryolagersystem probenarchivierungssystem biorepositorium kryodewar dewargefäß probenlagerdewar cellbankingsystem zellbanklagersystem kryolagerfürzelllinien lagerungvonsäugerzellen zellkryolager zelllinienrepository stammzellkryolager stammzellbiobank dnabiobank rnaprobenlager biomateriallager cellbanking kryokonservierung hochdichtekryogenelagerplattform Specifications LN2 capacity in gas storage (liters): 24 / 90 LN2 capacity in liquid storage (liters): 480 Empty weight (kg): 230 / 323 Neck opening (mm): 800 Outside diameter (mm): 900 Total height (mm): 1205
AC SYS 480 with frame for liquid phase storage - MDR + CryoFill 2.0 PD MDR with Installation + Rack Storage System for AC SYS 480 R2
1. Please choose: STORAGE OBJECT & PHASE
VIALS 2ml IN LIQUID PHASE, VIALS 2ml IN GAS PHASE, STRAWS IN LIQUID PHASE, STRAWS IN GAS PHASE
2. Please choose: AUTOMATIC FILLING SYSTEM
WITH RACK STORAGE SYSTEM, WITH CANISTER STORAGE SYSTEM, WITHOUT STORAGE SYSTEM
AC SYS 480 without frame for liquid phase storage - MDR + CryoFill 2.0 PD MDR with Installation + Canister Storage System for AC SYS 480 C2
AC SYS 480 with frame for liquid phase storage - MDR + CryoFill 2.0 PD MDR with Installation
AC SYS 480 with frame for liquid phase storage - MDR + Rack Storage System for AC SYS 480 R2
AC SYS 480 without frame for liquid phase storage - MDR + Canister Storage System for AC SYS 480 C2
AC SYS 480 with frame for liquid phase storage - MDR
AC SYS 480 with frame for gas phase storage - MDR + CryoFill 2.0 PD MDR with Installation + Rack Storage System for AC SYS 480 R1
AC SYS 480 with frame for gas phase storage - MDR + CryoFill 2.0 PD MDR with Installation + Canister Storage System for AC SYS 480 C1
AC SYS 480 with frame for gas phase storage - MDR + CryoFill 2.0 PD MDR with Installation
AC SYS 480 with frame for gas phase storage - MDR + Rack Storage System for AC SYS 480 R1
AC SYS 480 with frame for gas phase strorage - MDR + Canister Storage System for AC SYS 480 C1
AC SYS 480 with frame for gas phase storage - MDR
AC SYS 480 without frame for liquid phase storage - MDR + CryoFill 2.0 PD MDR with Installation
AC SYS 480 without frame for liquid phase storage - MDR
AC SYS 480 with frame for gas phase storage - MDR + Canister Storage System for AC SYS 480 C1