Cooling Biochemical Incubator Capacity: 103L, 152L, 253L, 412L, 722L, 1022L Temperature Range: 0~65℃, -10~65℃ Circulation Mode: Forced Convection (Balanced Type) The new diffusion-type air duct circulation system improves temperature uniformity Application Cooling Biochemical Incubators are suitable for the cultivation and preservation of bacteria and other microorganisms, plant breeding and seedling experiments, as well as water quality analysis and BOD determination. Performance Advantages The frost-free design effectively avoids temperature drift caused by the absence of a defrosting function, or significant fluctuations in temperature during traditional defrosting cycles. It maintains excellent temperature stability even during long-term experiments. The circulating fan has three adjustable speeds (high, medium, and low) to prevent sample evaporation due to excessive airflow during experiments. The new diffusion-type air duct circulation system improves temperature uniformity. The refrigeration system adopts hot gas bypass energy regulation technology to ensure efficient and safe operation of the compressor under high-temperature conditions. The programmable microcomputer fuzzy PID control technology, with 30 segments and 99 cycles, enables gradient and slope control of temperature rise and fall through the programmed settings of parameters such as multi-segment temperature, circulating wind speed, running time, and heating rate. Structural Features Wide-screen display: Convenient for setting programs and observing temperature, fan speed, running time, etc. It is equipped with temperature alarm function (indicator and audible prompt). Over-temperature protection device: Automatically interrupts operation when the preset temperature range is exceeded, featuring audible and visual alarms to ensure the safety of samples and the experimental environment. USB interface: Used to connect a printer or USB flash drive for real-time recording of temperature data during the experiment. Auxiliary lighting: Allows easy observation of the chamber interior in low-light environments; waterproof, resistant to high and low temperatures, and offers a long service life. Test hole: Convenient for monitoring; equipped with a silicone soft plug to prevent air convection. Outer casing: Made of high-quality cold-rolled steel plate with powder coating technology, featuring uniform thickness and a sleek, beautiful appearance. Inner Chamber and Shelves: Precision-crafted from SUS304 stainless steel; the shelves are adjustable to the desired height. Square steel base: Constructed from integrally welded rectangular square steel for high strength and durability, ensuring stability without vibration. Universal casters: Can be freely turned and locked; made of wear-resistant polyurethane. Specification Model ICB-B103YB ICB-B152YB ICB-B253YB ICB-B412YB ICB-B722YB ICB-B1022YB ICB-B103YL ICB-B152YL ICB-B253YL ICB-B412YL ICB-B722YL ICB-B1022YL Capacity (L) 103 152 253 412 722 1022 Circulation Mode Forced Convection (Balanced Type) Dimension (W*D*H) Internal (mm) 480*390*550 500*460*660 640*460*860 560*530*1390 980*530*1390 980*750*1390 External (mm) 645*700*1255 665*780*1360 800*780*1570 1025*810*1735 1445*810*1735 1445*1030*1735 Temperature Range (℃) YB Series: 0~65; YL Series: -10~65; Resolution (℃) 0.1 Fluctuation (℃) High temperature ±0.3℃, Low temperature ±0.5℃ High temperature ±0.5℃, Low temperature ±1℃ Uniformity (℃) ±0.5 ±1 Operating Ambient Temperature (℃) 5-30 Shelves (Standard) 2 3 3 4 4 4 Test Hole (mm) φ40 Electricity Voltage/Frequency AC 220V, 50Hz Consumption (W) 780 840 1200 1500 2100 2550 Safety Protection Refrigeration System Compressor overheating, over-current and over-pressure protection; condenser fan overheating protection. Others Main power phase sequence and phase loss protection; Leakage and short-circuit protection; Overload protection.
ICB-B103YB, ICB-B152YB, ICB-B253YB, ICB-B412YB, ICB-B722YB, ICB-B1022YB