IFM LMT100 - Point Level Sensor, Hygienic G1/2, IO-Link
IFM LMT100 Point Level Sensor — factory-set for water-based media, hygienic G 1/2 sealing cone, 2 outputs and IO-Link. capacitive point level detection for liquids, viscous media and powders, hygienic G 1/2 sealing-cone connection, factory-set for water-based media, 2 switching outputs and IO-Link, PEEK sensor tip, IP68/IP69K. Shipping & Returns Same-day dispatch across UAE Order before 2 PM (Sun–Thu). Next-day delivery to Dubai, Sharjah, Abu Dhabi. 2–4 days to other emirates. 7-day return window Sealed, unused items can be returned within 7 days for full refund or replacement. Compatibility-checked Our specialists review every bulk quote to ensure the part fits your application before you commit. MPN / SKU: LMT100 Also known as: IFM LMT100 level sensor / level switch. LMT100 Applications The IFM LMT100 point level sensor detects level in tanks and containers across the food, beverage and process industries. Its hygienic design and residue/foam suppression give reliable switching in liquid, viscous and bulk media, and the factory setting is optimised for water-based products. With two configurable outputs and IO-Link, it supports overflow and dry-run protection as well as media differentiation in CIP-cleaned plant. LMT100 Specifications Brand IFM Product Type Sensor for point level detection Product type code LMACE-A12E/QSKG/0/US Measuring principle Capacitive Factory setting Water-based media Media Liquid & viscous media; powder Tank pressure -1...40 bar Medium temperature -40...100 °C (150 °C short-time) Process connection G 1/2 sealing cone Operating voltage 18...30 V DC Outputs 2 x switching (PNP/NPN); IO-Link Protection IP 68; IP 69K Wetted material PEEK Weight 169.5 g Electrical connection M12 connector, A-coded, 4 contacts (gold-plated) LMT100 Datasheet & Manual Download Datasheet & ManualIFM LMT100 · PDF · Official IFM document LMT100 – Frequently Asked Questions (FAQ) What operating principle does the IFM LMT100 point level sensor use?The IFM LMT100 uses a capacitive measuring principle with an 11 mm probe to detect liquid and viscous media and powder; its factory setting is for hydrous media.What is the response time of the IFM LMT100?The IFM LMT100 has a response time of less than 0.5 s.What outputs does the IFM LMT100 provide?The IFM LMT100 has 2 digital outputs (switching signal and IO-Link), PNP/NPN, parameterisable normally open / normally closed, with 100 mA current rating and a max. voltage drop of 2.5 V DC.Does the IFM LMT100 support IO-Link?Yes, the IFM LMT100 supports IO-Link revision 1.1 with COM2 (38.4 kBaud) transmission per IEC 61131-9, with a minimum process cycle of 2.3 ms.What is the operating voltage and current consumption of the IFM LMT100?The IFM LMT100 operates on 18...30 V DC, draws less than 50 mA, and has reverse polarity protection.What is the IP rating and temperature range of the IFM LMT100?The IFM LMT100 is rated IP 68 / IP 69K, with an ambient range of -40...85 degrees C and a medium temperature of -40...100 degrees C (up to -40...150 degrees C short-term for 1 h).What are the wetted materials and process connection of the IFM LMT100?The IFM LMT100 uses stainless steel (316L/1.4404), PEEK, PEI and FKM, with a PEEK wetted surface (Ra < 0.8 um), and a G 1/2 sealing cone process connection.What pressure rating and connector does the IFM LMT100 have?The IFM LMT100 withstands tank pressure of -1...40 bar (-0.1...4 MPa) and uses an M12 A-coded 4-contact gold-plated connector.