hamamatsu Microchannel plate photomultiplier tube R5916U-52
hamamatsu Microchannel plate photomultiplier tube R5916U-52 Download Datasheet: hamamatsu Microchannel plate photomultiplier tube R5916U-52 55 mm dia., MCP-PMT, Bialkali photocathode (Effective area : 10 mm dia./Spectral response : 160 nm to 650 nm) Specifications Type MCP-PMT Tube Size Dia.55 mm Photocathode Area Shape Round Photocathode Area Size Dia.10 mm Wavelength (Short) 160 nm Wavelength (Long) 650 nm Wavelength (Peak) 400 nm Spectral Response Curve Code 403K Photocathode Material Bialkali Window Material Quartz Dynode Structure MCP Dynode Stages 2 [Max. Rating] Anode to Cathode Voltage -3400 V [Max. Rating] Average Anode Current 0.0001 mA Anode to Cathode Supply Voltage -3000 V [Cathode] Luminous Sensitivity Min. 20 μA/lm [Cathode] Luminous Sensitivity Typ. 45 μA/lm [Cathode] Radiant Sensitivity Typ. 45 mA/W [Anode] Luminous Sensitivity Typ. 9 A/lm [Anode] Gain Typ. 3.0 x 105 [Anode] Dark Current (after 30min.) Max. 0.5 nA [Time Response] Rise Time Typ. 0.18 ns [Time Response] Transit Time Typ. 1 ns [Time Response] Transit Time Spread Typ. 0.1 ns