Constant Temp. & Humidity Chamber Capacity: 173L, 288L, 400L, 516L Temperature Range: 0~65℃ Humidity Range: 40%RH ~ 95%RH (chamber temperature＞10°C) PID temperature controller, 24-segment program Application This product has a precise temperature and humidity control system, providing various environmental simulation conditions required for industrial research and biotechnology testing. It can be widely used for sterile testing, stability inspection, and testing of raw material performance, product packaging, product life, etc. in pharmaceuticals, textiles, food processing, and other industrial products. Features Outer casing material: Cold-rolled steel plate with powder coating; Inner chamber material: SUS304 stainless steel plate. Microcomputer intelligent PID temperature controller for precise temperature control; Equipped with parameter memory function, automatically resuming operation after power restoration. Uses SECOP refrigeration compressor, high efficiency and low consumption; uses CFC-free environmentally friendly refrigerant. The inner chamber is equipped with a fan to create micro-airflow circulation, improving the temperature uniformity inside the chamber. Equipped with a built-in water tank and automatic water replenishment interface. Automatically disconnects the power supply to the humidification heater when there is no water detected. The inner chamber adopts a rounded arc structure design for easy cleaning. Removable shelves with adjustable spacing. Equipped with casters for easy mobility. Standard φ35mm test hole. Standard USB data storage. (Optional) Independent external temperature protection device. (Optional) Micro thermal printer or dot matrix printer. Specification Model ICB-H180Y ICB-H290Y ICB-H400Y ICB-H520Y Capacity (L) 173 288 400 516 Circulation Mode Forced convection Dimension (W*D*H) Internal (mm) 550*450*700 600*600*800 800*550*910 800*710*910 External (mm) 700*840*1590 800*990*1690 1000*910*1800 1000*1070*1800 Temperature Range (℃) 0~65 Uniformity (℃) ±1 Fluctuation (℃) ±0.5 Display Accuracy 0.1 Control Accuracy ±0.1 Humidity Range 40%RH ~ 95%RH (chamber temperature＞10°C) Fluctuation ±3%RH Display Accuracy 1%RH Control Accuracy ±1%RH Humidification Method Built-in electric heating humidification water tank Controller PID temperature controller, 24-segment program Display LCD Shelves 3 3 3 3 Alarm Types Over-temperature alarm, under-temperature alarm, temperature probe failure alarm, humidity probe failure alarm Electricity Voltage/Frequency 1Ø, 220V/50Hz Consumption (W) 2000 2000 3000 3000 N.W. / G.W. (kg) 165/209 203/256 218/278 237/297
ICB-H180Y, ICB-H290Y, ICB-H400Y, ICB-H520Y