Pt1000 Surface Mounted RTD Temperature Sensor with Miniature Steel Threaded Fitting and Stripped Wire Leads
Pt1000 Surface Mounted RTD Temperature Sensor with Miniature Steel Threaded Fitting and Stripped Wire Leads RTD Sensor has a 1/4" Steel Hex fitting with a #8-32 UNC 2A .297" (7.5.mm) long Threaded Stud. The fitting transitions to PFA insulated lead wire with stripped ends. Temperature Operating range is -90 to 500°F (-70 to 260°C) Pt1000 (Platinum 1000 Ohm) Class A Thin Film Element encased in the Stainless Steel Fitting and secured with a high temperature epoxy. Lead Wire - 12 inches (1ft) or more of PFA (Plastic Fluoropolymer) insulated wire with stripped leads is provided for the connection to display or control instrumentation. Wire Construction a 3-conductor of 26 Gage Stranded (7/34) nickel plated copper. Each conductor is insulated with 500°F (260°F) rated PFA and twisted with an overall PFA jacket. Outside wire diameter is approx. .085". Color Code is 2 black and 1 red with a white overall jacket EvoSensors Part Number: P3A-THSF-THD-PK-0.31-PFXX-40-STWL with 40 Inch Long Wire Leads P3A-THSF-THD-PK-0.31-PFXX-12-STWL with 12 Inch (1ft) Long Wire Leads Consider the following Products to complete your Temperature Measurement System
12 Inch (1ft) Long Wire Leads, 40 Inch (1 meter or 3.3ft) Long Wire Leads, 80 Inch (2 meters or 6.6ft) Long Wire Leads, 120 Inch (3 meters or 10ft) Long Wire Leads