CONTEC08A Digital Upper Arm Blood Pressure Monitor with child cuff 18-26cm + adapter +PC Software
Introduction The Electronic Sphygmomanometer can be used to measure NIBP of human body. Features Fully automatic blood pressure measure The Electronic Sphygmomanometer stores the measure results of three users automatically, and up to 100 items for every user.It can satisfy blood pressure measure requirement of all your family. Three kinds of measure modes : adult, pediatric and neonatal.(Not available for OTC markets) By data review interface such as "data list","trend graph","big font", NIBP data is clear at a glance. With SpO2 measure function(with optional SpO2 probe).(Not available for US markets) Has the function of physiological alarm.It can be set the alarm limits,when blood pressure is higher than the high limit or lower than the low limit, the physiological alarm will occur.Alarm switch can be set. Screen displays prompt message when the power is low，and the device gives low power prompt sound.The prompt sound switch can be set. When there are factors which affect the measure in the process of measure and the device can't get the measure result,the device can display the corresponding error message. With short-time power storage function, when replacing the battery, the clock time function can't be affected. Supply two kinds of NIBP measure unit: mmHg / kPa Store measure results with date and time. High-definition color LCD display,supply English/Chinese interface, strong visibility. Function of automatic power-off. Performance NIBP Measure Method: Oscillometry Measure Mode: The upper arm measure Measure range: 0kPa（0mmHg）~ 38.67kPa（290mmHg） Resolution: 0.133kPa（1mmHg） Accuracy: ±0.4kPa（±3mmHg） NIBP pulse rate measure range: 40～240bpm SpO2(Not available for US markets) Measure range: 35%～100% Accuracy: ±2%, 70%～100% SpO2 Probe Pulse Rate: Measuring range:30bpm～250bpm Resolution:1bpm Display: 2.8" High-definition color LCD Power: Four "AA" Batteries/5V or 6V Power adapter Safety character:ClassⅡ Type BF Physics speciality Dimension: 130mm*110mm*80 mm Weight: 300g (No Batteries) Accessories Cuff for adult(22-32cm) Adapter User manual（GB only）on request for other Language USB data line