Standard Process Interface Cable for Xi 80 / Xi 410 / Xi 1M
Standard Process Interface (PIF) Cable for Ethernet-Based Xi Series Infrared Cameras Adds Analog and Digital Inputs and Outputs for Direct Process Integration 0 to 10 V Analog Input and 0/4 to 20 mA Analog / Alarm Output RS485 Serial Communication for Stable Data Transmission Over Long Distances Integrated Electronics and a 5-Outlet Terminal Block for Fast, Tool-Free Wiring Fully Programmable Through the Free PIX Connect Software Rugged IP67 / NEMA 4 Construction Optional High-Temperature Version Rated to 250 °C (482 °F) Compatible with the Xi 80, Xi 410 LT, and Xi 1M If you purchase a camera with the water cooled housing, you cannot use the cables that come with the unit. You must purchase them separately and the cable part number needs to end in CG4. The Standard Process Interface (PIF) cable turns an Ethernet-based Xi series thermal camera into a fully integrated part of your control system. It carries analog and digital signals in and out of the camera and adds RS485 serial communication, all terminating in a labeled terminal block that wires straight to your PLC, controller, or process equipment. If you need your camera to trigger on an external event, output a live temperature to a controller, or raise an alarm, this is the cable that makes it happen. One Cable for Analog, Digital, and RS485 The PIF cable provides a 0 to 10 V analog input, a 0/4 to 20 mA analog output that can also serve as an alarm output, and a 24 V digital input, plus RS485 serial communication. That combination lets the camera both receive commands from your process and send temperature or status data back to it. Integrated electronics handle the signal conditioning, and a 5-outlet terminal block with clear color coding (ground, digital input, power, analog input, and analog output) makes wiring quick and unambiguous. Reliable Signals Over Long Distances RS485 is built for the plant floor, carrying data reliably over long cable runs where other interfaces would degrade. That makes it easy to place the camera where the measurement needs to happen and route its data back to a control room or cabinet without losing signal integrity along the way. Programmable Through PIX Connect Every input and output is configured in the free PIX Connect software, so you decide exactly what each line does. Analog and digital inputs can set emissivity, apply ambient temperature compensation, supply a reference temperature, control the flag, trigger snapshots, recording, line scans, or the event grabber, and reset peak or valley hold. The analog output can broadcast the main measurement area, a specific measurement area, internal temperature, flag status, an alarm, or the center pixel as a direct output. You get precise, repeatable control without any custom hardware. Built for Harsh, Hot Environments The cable is rated IP67 / NEMA 4 against dust and water, with a PVC jacket and a standard operating range of 0 °C to 85 °C. For installations near ovens, furnaces, or other heat sources, an optional high-temperature version withstands ambient temperatures up to 250 °C (482 °F). A right-angle connector option and a version prepared for use with the Xi water-cooled housing are also available for tight or demanding mounting situations. Compatibility This cable is designed specifically for the Ethernet-based Xi series and is compatible with the Xi 80, Xi 410 LT, Xi 410 MT, and Xi 1M. It is not compatible with the PI series or with the Xi 400 and Xi 640 cameras, which use a different pinout. Need More Channels or Isolation? The Standard PIF cable covers the essentials: one analog input, one digital input, and one analog or alarm output. For applications that require several independent outputs, additional inputs, galvanic isolation, and a fail-safe relay, a separate Industrial Process Interface is available. If you are not sure which fits your application, our engineers can help you choose. [show_stocked_pn] [to_order_table] To Order click on the model no.(or scroll to the top of page and select your configuration) then click the ADD TO CART button Model No. Cable Length Operating Temp Version ACXIIOCB1 1 m 85 °C Standard ACXIIOCB1CG4 1 m 85 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB1RC 1 m 85 °C Right Angle Connector ACXIIOCB5 5 m 85 °C Standard ACXIIOCB5CG4 5 m 85 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB5RC 5 m 85 °C Right Angle Connector ACXIIOCB5H 5 m 180 °C Standard ACXIIOCB5HCG4 5 m 180 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB5H2 5 m 250 °C Standard ACXIIOCB5H2CG4 5 m 250 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB10 10 m 85 °C Standard ACXIIOCB10CG4 10 m 85 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB10RC 10 m 85 °C Right Angle Connector ACXIIOCB10H 10 m 180 °C Standard ACXIIOCB10HCG4 10 m 180 °C Prepared for Usage with Xi Water Cooled Housing ACXIIOCB10H2 10 m 250 °C Standard ACXIIOCB10H2CG4 10 m 250 °C Prepared for Usage with Xi Water Cooled Housing [to_order_table] SPECIFICATIONS Cable Type: I/O process interface with integrated electronics Connector 1: Terminal block with 5 outlets Connector 2: 5-pin cable plug (to camera) Color Coding: Ground (brown), Digital Input (gray), Power (white), Analog Input (green), Analog Output (yellow) Analog Input: 0 to 10 V Analog / Alarm Output: 0/4 to 20 mA Digital Input: 24 V Serial Communication: RS485 Outputs / Inputs: 1x analog / alarm output, 1x analog input, 1x digital input Power Supply: 5 to 24 V DC Cable Length: 5 m standard; 1 m and 10 m also available Jacket Material: PVC, black Operating Temperature: 0 °C to 85 °C (high-temperature version to 250 °C / 482 °F) Storage Temperature: 0 °C to 85 °C Relative Humidity: 10 to 95 %, non-condensing Protection Class: IP67 / NEMA 4 Programming: via PIX Connect software Compatible Cameras: Xi 80, Xi 410 LT, Xi 410 MT, Xi 1M Scope of Supply / What comes with your order Standard Process Interface (PIF) cable with integrated electronics, a 5-outlet terminal block, and a 5-pin camera plug, in the length and version selected at the time of order.
Standard, Prepared for Usage with Xi Water Cooled Housing, Right Angle Connector