Sample Volume Fast Analysis Connectivity 9 Milk Parameters Easy Operation Memory for 50 measurements More Info The milk analyzers from Master series are high – quality devices for measurement of nine milk parameters. Precise analysis of the components can be done for 60sec./40sec./30sec. Master LM2 can be equipped with second peristaltic pump for automatic cleaning. Master LM2 is comfortable and reliable milk analyzer with sophisticated sensor with long life and better precision. Options by customer request – conductivity, inbuilt printer, USB flash memory stick, second peristaltic pump for automatic cleaning, ph.Three calibration channels are available. Standard calibrations for milk analyzer Master LM2 are for Cow milk, Sheep milk, UHT. The analyzer can be calibrated also for: Buffalo milk, Pasteurized milk, Goat milk, Mixed milk(cow 50% buffalo 50%), Skimmed milk, Whey and other types of milk and milk derivatives. Measuring Ranges Fat 0℅ -25℅ ±0.1℅ Solids-Non-Fat (SNF) 3℅ - 40℅ ±0.15℅ Density 1000 – 1160 kg/m3 ±0.3kg/m3 Protein 2℅ - 15℅ ±0.15℅ Lactose 0.01℅ - 20℅ ±0.2℅ Salts 0.4℅ - 4℅ ±0.05℅ Add Water 0℅ - 70℅ ±3℅ Freezing Point -0.4˚C to -0.7˚C ±0.005˚C Sample Temperature 1˚C-42˚C ±1˚C *Conductivity 2 – 14mS/cm ±0.1mS/cm *pH 0 - 14 ±0.05 INSTRUCTION MANUAL BROCHURE SOFTWARE Ultrasonic Technology Very small quantity of milk needed Memory for 50 measurements pH (option) Self calibration without computer Low power consumption 1 x RS232 connection 1 x USB connection to PC) ESC POS Printer Support Built-in keypad Large graphic display Peristaltic pump One year full waranty Clock (time and date) No acids or other chemicals are used