Ritter Midmark M7 SpeedClave – (Pre-Owned)
Ritter Midmark M7 Manual Autoclave, Pre-Owned The Midmark Ritter M7 SpeedClave is a reliable manual steam sterilizer designed for medical, dental and other clinical applications. The M7 is manually operated and features simple mechanical controls for dependable sterilization. Its compact countertop design accommodates wrapped and unwrapped instruments, packs and glassware, with a fast unwrapped cycle when the sterilizer is already hot. DETAILS SPECS DOCUMENTATION Manual Steam Sterilizer Simple and dependable manual controls Fast Unwrapped Cycle – Starting hot, sterilizes unwrapped instruments in approximately 9 minutes Adjustable sterilization temperature from 250°F (121°C) to 270°F (132°C) Temperature gauge continuously displays the chamber temperature Adjustable cycle timer from 0–30 minutes Timer automatically shuts the heater off at the end of the selected sterilization cycle Audible signal indicates when the cycle is complete Safety door prevents opening while the chamber is pressurized Low-water cut-out helps protect the heating element if there is insufficient water Pilot light indicates when the heating element is operating Self-contained water reservoir Sterilizes wrapped instruments, unwrapped instruments, packs and glassware Three instrument trays included Compact countertop design No Built-In Printer Chamber Size: 7.5″ x 15″ Overall Dimensions (L x W x H): 19″ x 14″ x 13″ Trays: 3 – Two Large and One Small Small Tray Size (L x W x H): 12″ x 4-1/8″ x 7/8″ Large Tray Size (L x W x H): 12″ x 5-5/8″ x 7/8″ Standard Unwrapped Cycle Time: Approximately 9 Minutes Starting Hot Unwrapped Instruments: 270°F / 132°C – 3 Min Wrapped Instruments: 270°F / 132°C – 15 Min Packs: 250°F / 121°C – 30 Min Voltage: 115V Frequency: 60 Hz Power: 1300W Current: 10A Midmark M7 Speedclave Brochure (PDF Download) Midmark M7 Speedclave Operators Manual (PDF Download)