سنسور هواشناسی اولتراسونیک Lambrecht مدل u[sonic]WS6
Professional Line u[sonic]WS6 Ident-No. 00.16480.000000 Measuring range wind direction 0…359.9 ° Measuring range wind speed 0…65 m/s Air temperature measuring range -40…+70 °C Measuring range rel. humidity 0…100 % RH Air pressure measuring range 300…1100 mbar Wind direction accuracy < 2° (> 1 m/s) RMSE Wind speed accuracy ± 0.2 m/s RMSE (v < 10 m/s); ± 2% RMSE (10 m/s < v < 65 m/s) Air temperature accuracy ± 0.1 K (0…60 °C); ± 0.2 K (-40…0 °C) Relative humidity accuracy Typical ± 1.5% (0…80%) RH; ± 2% (>80%) RH Air pressure accuracy ± 0.5 mbar Wind direction resolution 0.1° Wind speed resolution 0.1 m/s Air temperature resolution 0.1 °C Resolution rel. humidity 0.1 % Air pressure resolution 0.1 mbar Response threshold 0.1 m/s Output RS-485; RS-422 (optional; please specify when ordering: Part No.: 97.16470.000422 RS-422 output configuration) Protocols NMEA 0183 Modbus RTU (optional; please specify when ordering: ID no.: 97.16470.000001 Modbus configuration) SDI-12 (optional; please specify when ordering: Id no.: 97.16470.000002 Configuration SDI-12) Further protocols on request Interface RS-485/422; SDI-12 (optional) Measuring rate 0.1…10 Hz Operating conditions -40…+70 °C (with heating: -50…+70 °C); 0…100% RH Supply voltage Without heating: 6…60 VDC; with heating: 24 V AC/DC ± 20% Power consumption Sensor: typically 50 mA at 24 VDC; with heating: max. 10 A at 24 V AC/DC Heating data Factory configurable: 60 W / 120 W / 240 W (standard) Dimensions Ø 199 mm; height 238 mm Housing Seawater-resistant aluminum Protection class IP66; IP67 Weight Approx. 3.1 kg Standards and norms Low voltage standard: 72/23 EEC Protection class: DIN EN 60529 DIN EN 50121-4:2016 Salt spray: EN 60945 IEC 61724-1 Cold test Ad according to DIN EN 60068-2-1 (01/2008) Condensation test CH according to ISO 6270-2 (09/2005) and DIN EN ISO 12944-6 (07/1998), category C4 Salt spray test according to DIN EN ISO 7253 (04/2002) and DIN EN ISO 12944-6 (07/1998), category C4 EMC standards / Electrical safety DIN EN 60945; DIN EN 61000-4-2, 3, 4, 6, 11 Options (order separately) 00.16480.000200 u[sonic]WS6 with bayonet plug 32.16420.066100 Cable 10 m; with 12-pin bayonet plug, assembled Accessories (order separately) 32.16470.060000 Sensor cable, 15 m, 8-pin M16 plug