Autoclave SA-260MA – 24 Litre Autoclave SA-260MA – 24 Litre Specifications: Microprocessor control system: Fully automatic control from pre-vacuum to dry and 24 programs for your selection Pre-heating by dry heater ensuring chamber temperature balance and perfect drying function Special LIQUID sterilization program: it is the special function for laboratory and scientific use Special PRION sterilization program: it is the special program for PRION sterilization Special BOWIE-DICK test program: the users can identify the perfect sterilization function with this program An “ERROR” indicator, to help the users and engineers to find the function problem automatically MODEL : SA-260MA (B-CLASS) Overall Dimensions 655mm x 533mm x 442mm Chamber Dimension 260mm x 450mm Capacity 24 l Temperature / Pressure Unwrapped / Wrapped: 121 – 134°C ; 2.2kg per cm² Liquid : 121°C ; 1.2kg per cm² Control System Pressure control Power Consumption 220 V: 1.8kW ; 8.3Amp / 240 V: 2.1kW ; 8.8Amp Auto-fill Water System Yes Double Water Level Check Device Yes: One for water tank ; one for chamber water level check Auto Drying Yes: 0, 15, 30, 35 or 40 minutes Pre-Vacuum Yes Dry-Vacuum Yes Indicators / Temperature Display Yes / LED Sterilization Temp / Time Wrapped: 121°C – 30 min; 134°C – 15 min Unwrapped: 121°C – 15 min; 134°C – 4 min Standard Accessories Sterilization box x 1 ; Tray x 1 Optional Accessories Tray frame x 1 ; Tray holder x 1 ; Tray x 3 ; Spring x 2 Safety Devices ü Temperature control switch ü Pressure door auto-lock device ü Emergency key – cuts power supply & exhausts chamber pressure ü Chamber pressure gauge + pressure & steam pressure safety valve ü Door close indication ü Low water alarm and indication ü Over-heat protector ü Safety valve ü Electronic circuit safety system ü Low water indicator & shutout device ü Sterilization completion indicator Quality Standard CE-MDD