Tuttnauer T-Edge 10B & 11B – Class B Autoclave
Tuttnauer T-Edge 10B & 11B Class B Pre-Post Vacuum Sterilization Optimal chamber capacity Convenient and easy loading with door opening to 110º Assorted water filling and drainage methods (top, front or automatic) Range of automatic one-button programs Remote Wi-Fi monitoring via smartphone or tablet DETAILS SPECIFICATIONS DOCUMENTATION Fast Cycles: The T-Edge 10B completes an unwrapped cycle in approximately 20 minutes and a wrapped cycle in approximately 39 minutes. The T-Edge 11B completes an unwrapped cycle in approximately 26 minutes and a wrapped cycle in approximately 49 minutes. Cycle times include drying. Class B Sterilization: Advanced pre-vacuum and post-vacuum technology provides reliable sterilization and drying for wrapped, unwrapped, hollow and porous loads. Optimal Chamber Capacity: Utilizes more than 75% of the chamber capacity with five extra-long trays. Convenient Loading: Door opens to 110° for smooth, unobstructed loading. Remote Wi-Fi Monitoring: Track and monitor your autoclave using a smartphone or tablet. Advanced Documentation: Multiple documentation and traceability options, including barcode integration. User-Friendly Touchscreen Control Panel: Displays programs, cycle times, water levels and operating information. Details: T-Edge-10B T-Edge-11B Chamber Size: 10″ (Ø) x 18″ (D) 11″ (Ø) x 18″ (D) Volume: 23L / 6 Gal 27 Liters / 7 Gal Usable Capacity: ~17L (75%) ~20.5L (75%) Overall Dimensions: 22.8″ (D) x 18.9″ (W) x 19.7″ (H) 22.8″ (D) x 19.7″ (W) x 19.7″ (H) Max Trays: 5 Trays 5 Trays Tray Dimensions: 16.6″ (D) x 7.4″ (W) x 0.8″ (H) 16.6″ (D) x 8.1″ (W) x 0.8″ (H) Weight: 117 lbs. 124 lbs. Water Filling: Top, front or automatic Top, front or automatic Connectivity & Compliance Digital Connections: 1 Ethernet port, 2 USB ports and Wi-Fi Automatic Connections: Integrated automatic water-filling and drain connections Electrical Safety: EN/IEC 61010-2-40 and IEC 61010-1 Electromagnetic Compatibility: EN 61326-1 Functional Tests: EN 13060 and ANSI/AAMI ST55 Software Validation: EN 62304 and FDA guidance Usability Testing: IEC 62366-1 Quality Management: ISO 13485 Pressure Vessel Compliance: ASME Code Section I and Section VIII, Division I Health Canada Licence: T-Edge 10 and T-Edge 11 are listed under active medical-device licence No. 5836 Designed for efficiency and compliance, the T-Edge 10B and T-Edge 11B combine advanced Class B sterilization with high-capacity chambers, user-friendly operation and advanced connectivity. Tuttnauer T-Edge 10B Brochure (PDF Download) Tuttnauer T-Edge 11B Brochure (PDF Download) Tuttnauer T-Edge 10B Manual (PDF Download) Tuttnauer T-Edge 11B Manual (PDF Download) Tuttnauer T-Edge 10B Installation Guide (PDF Download) Tuttnauer T-Edge 11B Installation Guide (PDF Download) Tuttnauer T-Edge 10B Quick Start Guide (PDF Download) Tuttnauer T-Edge 11B Quick Start Guide (PDF Download)