Labconco FreeZone 4.5L -84° C Benchtop FreezeDryer with PTFE coil and collector 115V, 60Hz (710402000)
Labconco FreeZone 4.5L -84° C Benchtop FreezeDryer with PTFE coil and collector 115V, 60Hz (710402000) Manufacturer: Labconco Model: FreeZone 4.5L Part: 710402000 Condition: New Estimated lead time: ships in 6-7 weeks from order and payment. Warranty: 1 year manufacturer's warranty Includes: The Labconco 710402000 FreeZone 4.5L -84C Benchtop Freeze Dryer with PTFE coil and collector 115V, 60Hz Vacuum Hose Drain Hose Hose Clamp x 2 Baffle Power Cord Operator's Guide Manufacturer's Documentation 1 Year Manufacturer's Warranty Features Upright stainless steel collector coil capable of holding from 2.5L to 8L of ice before defrosting. Acrylic collector lid, 3/4" thick, has neoprene gasket and 3" diameter port for connection of drying accessories (sold separately). Compact benchtop design with a small footprint. Brushed stainless steel and glacier white, powder-coated steel exterior with blue accents. Full color, 5" capacitive touchscreen with Lyo-Works™ OS: Real time display of collector temperature, vacuum level, and optional End-Zone™ end point detection system. Start–up mode for collector cool-down and vacuum pull-down (automatic or manual). Operating mode (standby or run). Vacuum set point, alerts, diagnostic tests, user manual, defrost control, and optional drying accessory parameters. Data logging stores and displays data in table or chart format, and may be exported via USB or Ethernet. Settings include language (English, Spanish, German, French, Italian, Chinese, Japanese, Portuguese), time, time zone, temperature (° C or ° F), and vacuum (mBar, Torr, or Pa). Alerts are programmable to send via Ethernet connection to user-entered email addresses. Display settings include brightness, touch sounds (on/off), and screen time out in minutes (on/off). Maintenance alert settings include vacuum pump oil change, refrigeration cleaning, total operating hours and line voltage offset. Level-specific security lock-out prevents changes to programs and other system parameters. Hot gas defrost with auto shut off at +60° C (+140° F). On/standby switch. Moisture sensor that prevents refrigeration or vacuum start-up when moisture is detected in the drain line. Vacuum control valve that maintains vacuum level set point. Vacuum break valve that bleeds air into the system when power to the freeze dryer or vacuum pump is shut off. If a power outage less than approximately 5 minutes occurs, the freeze dryer will restart and the refrigeration and vacuum system will resume operation once power is restored. If the power failure is more than approximately 5 minutes and the collector warms above safe limits, the freeze dryer will not automatically restart. Side-mounted USB port and rear-mounted Ethernet connection. Side-mounted, quick-disconnect drain hose fitting and lanyard with hose connector. 18", 1/4" ID silicone tubing and clamp included. Rear-mounted electrical receptacle for vacuum pump connection. 3-wire electrical cord with plug. 3/4" OD vacuum connection, 50" of 3/4" ID vacuum hose and two clamps. Specifications Weight 197.0 Electrical 115V, 60 Hz, 16A Collector Temperature -84° C Ice Holding Capacity 4.5L Plug Type North America, 115V Style Benchtop Vacuum Pump Displacement Required 98 L/min or Larger Displacement Conformance CAN/CSA C22.2, UL Required Accessories (Not Included) Vacuum pump with a displacement of at least 98 L/min, 0.02 mBar ultimate pressure and fitting suitable for 3/4" ID vacuum hose. Freeze Dry Glassware, Small Tray Dryer, or other drying accessory