Industrial Process Interface for Xi Series Cameras
Industrial Process Interface (PIF) for USB-Native Xi Series Infrared Cameras (ACPIPIFMA) Expands Camera I/O - Three Analog/Alarm Outputs, Two Analog Inputs, and One Digital Input 500 VAC Galvanic Isolation Between Camera and Process for Safety and Noise Immunity Integrated Fail-Safe Monitoring - Triggers an Alarm on Cable Interruption or Software Shutdown Programmable Functions: Emissivity, Ambient Compensation, Frame Sync, Alarms, and Triggered Recording Rugged IP65 / NEMA 4 Aluminum Housing - Optional PCB-Only Version for Custom Integration High-Temperature Cable Options Rated to 180 °C or 250 °C; Lengths from 25 cm to 20 m The Industrial Process Interface (PIF) connects your USB-native Xi series infrared camera to the surrounding control system, turning live thermal data into the analog and digital signals that PLCs, alarms, and process equipment understand. Instead of relying on a directly attached PC, the PIF lets the camera output temperatures, statuses, and alarms - and accept external setpoints - over standard industrial I/O, making it the link between thermal measurement and automated process control. How It Differs from the Standard Process Interface Cable The standard process interface (PIF) cable for the Xi cameras (comes with the camera) that provides basic, non-isolated I/O - a single analog output (0-10 V), one analog input, and one digital input on a bare cable terminating in a 5-pin terminal block. The industrial PIF is a substantial step up. It expands the I/O to three analog/alarm outputs plus three switching alarm relays, two analog inputs, and a digital input, and its analog outputs are 0/4-20 mA current loops rather than 0-10 V - far better suited to long cable runs and electrically noisy plants. Most importantly, it adds 500 V AC galvanic isolation between the camera and the process, packages the electronics in a rugged, mountable enclosure with status LEDs, and continuously monitors for faults with built-in fail-safe alarms. The standard cable is fine for simple bench or light-duty signaling; the industrial PIF is built for permanent integration into industrial control systems. Expanded Analog and Digital I/O The interface provides three analog/alarm outputs, two analog inputs, and one digital input. Outputs (0/4-20 mA, or 0-30 V / 400 mA alarm relays) can transmit values such as the main measurement-area temperature, internal temperature, frame sync, alarm, and fail-safe status. Inputs (0-10 V analog, 24 V digital) let the process feed the camera parameters like emissivity, ambient temperature compensation, and reference temperature, or trigger snapshots, recordings, line-scan capture, and peak-/valley-hold resets. 500 VAC Isolation and Fail-Safe Protection A key benefit of the PIF is 500 V AC RMS galvanic isolation between the infrared camera and the process, protecting the camera and the connected PC from the voltage spikes and ground loops common around industrial machinery. Integrated fail-safe mechanisms continuously monitor the connection: if the cable is interrupted or the software shuts down, the PIF triggers an alarm rather than failing silently. On-board LEDs - two green for power and fail-safe, three red for alarm-relay status - give at-a-glance diagnostics. Compatibility The industrial PIF is designed for USB-native Optris Xi series infrared cameras, including the Xi 80, Xi 410 LT, Xi 410 MT, and Xi 1M. It connects to the camera's process-interface connection and is powered from a 5-24 VDC supply. If you are unsure whether your camera uses the USB process interface, confirm the model before ordering. Built for Industrial Environments Housed in a rugged aluminum enclosure measuring 122 x 70 x 27 mm, the standard PIF carries an IP65 / NEMA 4 rating and operates from -30 °C to 85 °C, with vibration and shock resistance tested to IEC 60068 standards. The connecting cable is available in lengths from 25 cm to 20 m, with high-temperature versions rated to 180 °C or 250 °C for hot installations, plus water-cooling-ready and right-angle-connector assembly options. A PCB-only version is available for customers integrating the board into their own housing. Downloads Industrial Process Interface Data Sheet Technical Note for Xi Models SPECIFICATIONS Compatibility: USB-native Xi series cameras (Xi 80, Xi 410 LT, Xi 410 MT, Xi 1M) Part Number: ACPIPIFMA Outputs: 3x analog / alarm outputs (0/4-20 mA); 3x alarm relays (0-30 V / 400 mA) Inputs: 2x analog inputs (0-10 V); 1x digital input (24 V) Isolation: 500 V AC RMS between infrared camera and process Power Supply: 5 … 24 VDC LED Indicators: 2x green (power, fail-safe); 3x red (alarm-relay status) Environmental Rating: IP65 (NEMA 4) Operating Temperature: -30 °C … 85 °C Storage Temperature: -30 °C … 85 °C Relative Humidity: 10 - 95 %, non-condensing Vibration: IEC 60068-2-6 (sinusoidal), IEC 60068-2-64 (broadband noise) Shock: IEC 60068-2-27 (25 G and 50 G) Housing Size: 122 x 70 x 27 mm Weight: 610 g (with 5 m cable) Cable Lengths: 25 cm, 5 m, 10 m, 20 m (high-temperature versions to 180 °C / 250 °C)
Standard, Prepared for Water Cooling Accessory, Right Angle Connector