SPC-QC-106 Five-Channel Time-Correlated Single Photon Counting TCSPC-FLIM Module/ six Channel Time-Tagger Module
Description Tech Specs Download PDF Item Description The SPC-QC-106 TCSPC / FLIM module five parallel TCSPC / FLIM channels with a common reference channel on a single PCI-express board. Alternatively, the module can be operated with six absolute photon timing channels. The module features high temporal and spatial resolution, high peak count rate, and extraordinarily high timing stability. The SPC-QC-106 offers the usual modes for recording temporal waveforms of optical signals, sequential recording multi-wavelength recording, time- and parameter-tag recording, FLIM, spatial and temporal mosaic FLIM, triggered accumulation of fast time series of curves and images, and simultaneous FLIM / PLIM. Six Channel TCSPC/FLIM Module Five Parallel TCSPC / FLIM Channels, One Synchronization / Reference Channel or Six Parallel Absolute Timing Channels Time Channel Width Down to 4 ps IRF Width < 38 ps FWHM Timing Jitter < 17 ps RMS Excellent Timing Stability: Timing Drift over 10 Minutes <5 ps RM Measurement Software Included The SPCM operating and measurement software is included with all SPC series modules. SPCM provides online calculation and display (2D, 3D) of data (decay curves, FLIM, FCS, FCCS) acquired in multiple operation modes. SPCM software undergoes active continuous development. SPCM receives frequent updates with new features and bug fixes. Product Technical Specs Summary product specifications are below View and download the detailed product sheet at the Download PDF tab Please contact us with any questions for or assistance via our contact request page or at 617-566-3821 Click the PDF tab above to see the data sheet Becker & Hickl SPC-QC-106 web page