About Compatible With Analytical Industries [PSR-11-915] Datascope [Multinex Analyzer] GE (Datex-Ohmeda) [0237-2034-700 Excel Modulus,5250 RGM,5150 Handheld Monitor] GE (Ohmeda) [5120, 5100, 5250 RGM, 4700 Oxicap, 7800, 5125, 5150, Handheld Monitor, 0237-2034-700, Excel Modulus] Hewlett Packard [15201A] Mercury Medical [10-103-07] Pacifitech [PT-10] Teledyne [R15 #C-41131] Teleflex (Hudson) [5556] VTI [100352, 101272, 100352] Specifications Range of Measurement: 0 to 100% Output Range: 14 to 20 mV Zero Offset: ≤ 0.30 mV Response Time: ≤ 17 seconds Linearity: ≤ 3% full scale Expected Lifetime: > 500,000 %O2 hours Temperature Compensation: none Operational Temperature Range: 10˚ to 40˚C Storage Temperature Range: -20˚ to 50˚C Humidity Range: 0% to 90% RH, non-condensing Ambient Pressure Range: 700-1250 mbar Interference: per EN ISO 80601-2-55 Electrical Interface: gold plated slip rings