Lovibond LC 10 Pure water conductivity cell for SD 320 / SD 325 with 1 m cable
Precise 2-pole stainless steel electrodes (K ≈ 0.1 cm-1) for the determination of low conductivities (≤ 200 µS/cm, also suitable for ultrapure water) Elimination of influences by atmospheric CO2 via a flow cell suitable for 12 mm probes Automatic temperature compensation (NTC 10 kΩ) The SD conductivity cell type LC 10 has been developed for conductivities below 200 µS/cm. With the use of two sensitive stainless-steel electrodes even the conductivity of ultrapure water can be determined easily and with highest accuracy. The integrated flow cell made of glass eliminates fluctuations in measured values caused by the introduction of CO2 from the air. The built-in NTC 10 kΩ thermistor provides the necessary automatic temperature compensation. Technical Data Measuring Range < 200 µS/cm Operating Temperature -5 - 80 °C Sensor Material Stainless Steel Sensor Type 2-Pole Integrated Temperature Sensor NTC 10 kΩ Outer Material Stainless Steel Cell Constante K ≈ 0.1 cm-1 Length 75 mm Diameter 12 mm Cable Length 1 m