Dansons PL350 Compact Electric Patient Lift – 350 lb Capacity
Dansons PL350 A full-body patient lift that fits the room you actually have The PL350 is Dansons Medical's compact electric patient lift: a 350 lb lifting capacity, a 17.5″ to 64.7″ lift range, and a base that closes to 23″ wide so it passes through standard doorways and works in bedrooms and bathrooms that larger lifts cannot reach. Electric raising and lowering means the caregiver presses a button instead of pumping a handle — the difference between a transfer that strains the back and one that doesn't. The base opens on foot pedals so both hands stay on the client, and a universal mesh sling with head support is included in the size you choose. Who it's for Home care households, family caregivers, and nursing or residential facilities moving clients between bed, chair, commode and wheelchair. Built around a 350 lb working capacity, which covers most non-bariatric transfers. Control Electric powered raising and lowering, with foot pedals opening and closing the base legs so the caregiver keeps both hands on the client. One person can complete a transfer without a second pair of hands. Where it fits A 23″ closed base clears standard interior doorways and opens to 33.9″ around a bed, chair or commode. Low-profile dual-wheel casters roll on carpet and uneven flooring rather than fighting it. Features 01 · Reach & Footprint 350 lb, 17.5″ to 64.7″, in a 23″ frame A 350 lb maximum lifting capacity across a range that reaches down to 17.5″. That low figure is what matters after a fall — it lets the lift reach a client on the floor, which is precisely the situation where a caregiver is most likely to injure themselves attempting it manually. At the top of its travel it reaches 64.7″, clearing most bed and chair heights. Most patient lifts are bought for a house that was never designed around one. A 23″ closed base clears standard interior doorways, so the lift travels between bedroom and bathroom rather than living in one room. Opened to 33.9″ it straddles a bed frame, armchair or commode, and maintenance-free dual-wheel casters at the front with braked casters at the rear roll over carpet and thresholds without snagging. 350 lb capacity 17.5″ – 64.7″ 23″ closed / 33.9″ open Floor pick-up capable 02 · Operation Electric lift, foot-pedal base Raising and lowering are electric powered, so the caregiver presses a button rather than working a hydraulic handle — steadier for the client and far easier on the operator's back across several transfers a day. The base legs open and close on foot pedals, which keeps both hands free and on the person being moved. Electric raise and lower Foot-pedal base opening Single-caregiver operation 03 · Power & Controls Battery powered, with an LCD that tells you where you stand A 24 VDC battery gives 15 full cycles per charge under full load, and an LCD on the control box shows battery status and warns on over-duty, so nobody is guessing whether there is enough charge left for the next transfer. The unit is rated IP54 against dust and splashing water, and runs at 16 dB under load — quiet enough not to wake a sleeping household. An emergency lowering device at the actuator shaft brings a suspended client down safely if the electronics fail. 15 cycles per charge IP54 rated Emergency lowering 04 · Sling & Attachment A full-body sling, included A universal mesh sling with head support ships with the lift in Small, Medium or Large — full-body, so the client is supported from head to toe rather than left with an unsupported neck through the transfer. The 4-point spreader bar takes 4, 6 and 8-strap slings, and 2-point and 6-point bars are available if your existing slings need one. Free sling included 4-point spreader bar Head support 05 · Safety & Working Limits Before you use it Maximum lifting capacity is 350 lb (160 kg). This is the limit for the whole system — the included slings are rated higher, but the lift's 350 lb rating always governs. The actuator runs a 10% duty cycle — a maximum of 2 minutes on followed by 18 minutes off. In a facility running back-to-back transfers, plan around it. Dansons directs that no transfer be attempted without prior approval from the person's nurse, and that the sling be chosen together with the patient's doctor, nurse or medical attendant. Read the owner's manual in full and practise the lift several times before transferring anyone. Inspect the sling for fraying, worn stitching or damaged straps before every use. Warranty runs 3 years on the lift frame and spreader bar, 2 years on the actuator, casters and pendant, and 6 months on reusable slings. It covers the original purchaser only and excludes misuse, improper installation, neglect and normal wear. This equipment should be used in line with guidance from the client's physician, occupational therapist or other qualified healthcare professional. Morelli Medical does not provide clinical assessment or prescribe equipment. Specifications Dansons PL350 Compact Electric Patient Lift Model / SKU PL350 · DM-PL350 Manufacturer Dansons Medical Product Type Full-body electric patient lift Maximum Lifting Capacity 350 lb / 160 kg Minimum Lifting Height 17.5″ Maximum Lifting Height 64.7″ Mast Height 45.0″ Base Legs Length 40.1″ Base External Width (Closed) 23″ Base External Width (Open) 33.9″ Lift Weight 77 lb / 35 kg assembled · heaviest detachable part 30.4 lb / 13.8 kg Shipping Weight 88 lb / 40 kg — ships by standard ground carrier, no freight required Lift System Electric actuator, powered raising and lowering Base Opening Foot pedal operated Spreader Bar 4-point (2-point and 6-point available separately) Casters 3″ dual front · 4″ rear with brake · maintenance free Emergency Lowering Mechanical emergency lowering device at the actuator shaft Warranty Lift frame and spreader bar 3 years · actuator 2 years · casters and pendant 2 years · reusable slings 6 months Availability Assembly required — basic tools supplied in the carton Control Unit Input Voltage 110–120 VAC 60 Hz / 220–240 VAC 50 Hz Output Voltage 24 VDC Battery Pack 24 VDC, 2.9 Ah Cycles Per Charge 15 full cycles under full load Duty Cycle 10% — maximum 2 minutes on, 18 minutes off Protection Class IP54 — protected against dust and splashing water Sound Level 16 dB under full load · 11 dB no load Display LCD battery status and over-duty warning Included Universal Mesh Sling — Sizing Size Shoulder circumference Best fits Sling rating Small 28″ – 45″ 85 – 140 lb 400 lb Medium 37″ – 57″ 140 – 200 lb 400 lb Large 47″ – 67″ 200 – 400 lb 450 lb Capacity note: the slings are rated to 400 lb and 450 lb, but the lift's 350 lb capacity is the limit for the whole system. Never load beyond 350 lb regardless of the sling rating. If you need a higher working capacity, call us on (866) 338-2838 and we will point you to the right model. Before first use: the manufacturer directs that no transfer be attempted without prior approval from the person's nurse, and that the sling be selected together with the patient's doctor, nurse or medical attendant. Read the owner's manual in full and practise the lift several times before transferring anyone. Warranty: covers defects in workmanship and product performance. It extends only to the original purchaser and is non-transferable, and claims require proof of sale and the serial number. Excluded: misuse, accidental or malicious damage, improper installation, use with incompatible parts, neglect, failure to maintain the product as the manual specifies, a removed or defaced serial number, and normal wear and tear. What's Included Dansons PL350 patient lift — base, mast, boom and actuator Power control unit with LCD battery and over-duty display Hand control Battery charger 4-point spreader bar 3″ dual front casters and 4″ rear casters with brake Detachable handle Emergency lowering mechanism Assembly tools (screwdriver supplied in carton) FREE universal mesh sling with head support in your selected size — Small, Medium or Large Owner's manual ↗ Product brochure ↗ Sold separately: 2-point and 6-point spreader bars, additional and replacement slings, spare batteries, replacement casters and other lift parts. Call us on (866) 338-2838 or email info@morellimedical.com if you need help sizing a sling or specifying a spare.
Dansons PL350 Compact Electric Patient Lift, Small - Dansons PL350 Compact Electric Patient Lift + Universal Mesh Sling w/ Head Support, Medium - Dansons PL350 Compact Electric Patient Lift + Universal Mesh Sling w/ Head Support, Large - Dansons PL350 Compact Electric Patient Lift + Universal Mesh Sling w/ Head Support