GX-3R Personal Multi-Gas Confined Space Monitor
Features: 3 Year warranty Simultaneous detection of 4 gases LEL, O2, H2S & CO Smallest / Lightest 4 gas monitor on the market (55mm W x 64mm H x 26mm D), approx. 100 g Simple 2-button operation Alarms: audible, visual, & vibration 3 user adjustable alarm levels Impact resistant body with removable rubber boot Large LCD Auto-backlit display 25 hours of operation (Lithium-ion battery) 40 hours in long-life mode Field replaceable sensors & filters Water and dust resistant design, IP68 Non-Compliance indicator Applications: Confined spaces Hazardous material Refineries / petrochemical Personal monitoring Utilities Oil and gas Water / wastewater plants Fire services Pharmaceuticals Construction GX-3R Four Gas Personal Monitor Overview R72.RA.C GX-3R 4 gas monitor EX- Type A (O2 / LEL / combo H2S/CO) with standard 240 VAC charger and calibration cup, calibrated for CH4, 3 year warranty The GX-3R one-to-four gas personal monitor picks up where the GX-2009 left off. It’s the world’s smallest four-gas personal monitor, weighing only 100 grams, and can comfortably be worn within the worker’s breathing zone (55mm W x 64mm H x 26mm D). It simultaneously monitors and displays LEL combustibles, O2, CO (Hydrogen compensated CO also available), and H2S. The Breathing Zone Optional Health and Safety defines the breathing zone as a 25cm radius around the worker’s nose and mouth and requires that worker exposure air samples be collected in the breathing zone. Air sampling filters may be attached to the collar or lapel. Traditionally, this means only a single gas monitor would be small enough for a worker to wear ‘In the breathing zone practically’. Now there is a four-gas monitor smaller than many single-gas monitors. Sensors This instrument represents the latest evolution of gas detection technology utilising a new generation of microsensors. The LEL sensor has a unique design with two active filaments in a single sensor to increase its resistance to silicone poisoning. If one filament gets poisoned, the second filament takes over. The new O2 sensor is electrochemical and has a 5-year lifespan. H2S and CO are now available in a combo microsensor. Non-compliance indicator Safety supervisors will appreciate the non-compliance indicator. The instrument flashes its 3 LED lights every 30 seconds, if the device has been exposed to a gas, has not been bump tested, or calibration is due. The 3 LED lights will continue to flash once every 30 seconds until the non-compliance condition has been resolved.