Professional Line PRO-WEA Ident-No. 00.14524.101040 Measuring ranges 0.5…60 m/s Accuracy ± 0.3 m/s ≤ 10 m/s; ± 0.5 m/s…60 m/s Resolution < 0.1 m/s Start-up value < 0.5 m/s Output 4…20 mA = 0…60 m/s Refresh rate 4 Hz (For the 4…20 mA output, the maximum load of 600 ohms must not be exceeded.) Areas of application Temperatures -40…+70 °C heated; wind speed max. gusts 100 m/s; relative humidity 0…100 % RH Supply voltage 24 VDC (20…28 VDC); 18 W heating; max. 800 mA. The heating in the sensor head prevents the moving parts from blocking under most climatic conditions. Measuring elements Aluminum; specially coated; 3-arm cup rotor Measuring principle Hall sensor array, contactless Housing Seawater-resistant aluminum; IP65 rating when installed vertically; M12 cable connector; stainless steel nut and lock washer Weight 0.35 kg Accessories (order separately) 32.14567.060000 Sensor cable with plug connection; Length: 12 m 32.14627.010000 Crossbar; Sensor spacing: 75 cm 32.14567.008000 Mast adapter; Diameter: 50 mm 32.14565.019000 Lightning protection rod